The nanoscale phase distinguishing of PCL-PB-PCL blended in epoxy resin by tapping mode atomic force microscopy
© Li et al; licensee Springer. 2012
Received: 9 October 2011
Accepted: 23 February 2012
Published: 23 February 2012
In this work, we investigated the bulk phase distinguishing of the poly(ε-caprolactone)-polybutadiene-poly(ε-caprolactone) (PCL-PB-PCL) triblock copolymer blended in epoxy resin by tapping mode atomic force microscopy (TM-AFM). We found that at a set-point amplitude ratio (rsp) less than or equal to 0.85, a clear phase contrast could be obtained using a probe with a force constant of 40 N/m. When rsp was decreased to 0.1 or less, the measured size of the PB-rich domain relatively shrank; however, the height images of the PB-rich domain would take reverse (translating from the original light to dark) at rsp = 0.85. Force-probe measurements were carried out on the phase-separated regions by TM-AFM. According to the phase shift angle vs. rsp curve, it could be concluded that the different force exerting on the epoxy matrix or on the PB-rich domain might result in the height and phase image reversion. Furthermore, the indentation depth vs. rsp plot showed that with large tapping force (lower rsp), the indentation depth for the PB-rich domain was nearly identical for the epoxy resin matrix.
Keywordstapping mode AFM PCL-PB-PCL phase image force-probe
Tapping mode atomic force microscopy (TM-AFM) has become a widely used technique to study the structures and properties of heterogeneous polymers at nanometer scale [1–9]. In a TM-AFM measurement, a cantilever is forced to oscillate with the probe tip at a given amplitude (A0). Then, the cantilever is brought close to the specimen and made to tap the surface with a given reduced set-point amplitude (Asp). The probe-sample interaction can introduce a phase shift in the vibration with respect to that of A0. So, TM-AFM measurement can obtain both height and phase images simultaneously. Height image can reflect the topographical and morphological structures, while phase images are sensitive to the physical and chemical properties of the studied material, such as stiffness, viscoelasticity, and chemical composition [3, 10–12].
However, the contrasts of the height and phase images sensitively depend on experimental conditions [1, 13–16] including cantilever force constant, tip shape, free amplitude A0, and set-point amplitude ratio (rsp) (equals to Asp/A0). This leads to difficulties in image interpretation for heterogeneous polymer samples. Thus, interpretation of the images has attracted considerable attention. To understand the results, a force-probe mode is performed on different materials [6, 13, 17]. In this mode, the amplitude and phase shift of the tapping cantilever are measured as a function of the varied tip-sample distance in order to study the indentation response of polymer surfaces in nanoscale. The tip penetration into the compliant sample is large, while very little penetration occurred on the stiff sample [18, 19]. If the matrix is more compliant compared with the domain regions or the domain size is much larger than the probe tip diameter (about 20 nm), the indentation difference is not affected by the tip size [1, 2, 6, 19]. In most of the reports, the investigated polymer samples were prepared by solution casting, so enrichment which occurred at the sample surfaces would affect the results except for the scan parameters [4, 6, 20, 21]. In this paper, we studied a new heterogeneous polymer which has a more compliant domain region with the size of about 20 nm by TM-AFM. The experiments were performed on the ultrathin section of the polymer to study the structure of the bulk material. To interpret the results, we carried out the force-probe measurement on the microdomains and the matrix.
Triblock copolymer of PCL-PB-PCL was synthesized, and the blend system of PCL-PB-PCL and epoxy resin was prepared according to the procedure reported in our previous work . The molecular weight of the PCL-PB-PCL was determined by hydrogen-1 nuclear magnetic resonance (1H NMR). It was found that Mn is 14,600 and the weight fraction of PB subchains in the PCL-PB-PCL is 35%. According to 1H NMR results, the content of PCL-PB-PCL in the polymer blends is about 10% by weight. The specimen section of the polymer blends was prepared using a microtome machine (ca. 70 nm in thickness) and used for TEM and TM-AFM examination.
The TEM measurement was performed with a high-resolution TEM instrument (JEM-2010, JEOL, Tokyo, Japan) at an acceleration voltage of 120 kV. The samples were stained with OsO4 to improve the image contrast. The stained specimen section was placed in a 200 mesh copper grid for observation.
TM-AFM measurements were performed with an AFM instrument (Nanoscope V, Veeco, Plainview, NY, USA) at ambient condition. The commercial Si cantilevers with two types of force constant (3 and 40 N/m) were used for tapping mode measurements. The height and phase images were recorded simultaneously using the AFM instrument. Tapping mode images were taken at the fundamental resonance frequency of the Si cantilever with an ultra-sharp tip (curvature radius approximately 10 nm). The typical scanning speed was 1 Hz for all measurements.
Structure of PCL-PB-PCL-blended epoxy resin film
Effect of cantilever stiffness on AFM image contrast
Under ambient condition, the contamination layer mainly composed of water is presented on the sample surface and leads to an attractive interaction between the cantilever and the surface water layer by capillary force . A weaker cantilever with smaller amplitude is easy to be controlled by this attractive force. So, in order to get clear contrast for a phase image, a stiffer cantilever and a larger amplitude should be used.
Effect of rspon AFM image contrast
It was reported that the relative image contrast of chemically different regions depended sensitively on rsp and probe tips [1, 14, 24, 25]. According to these works, in our research, a series of height and phase images were taken at different rsp (ranging from 0.95 to 0.1) using a stiffer tip with a force constant of 40 N/m.
Force-probe measurement on different domains
Figure 6a shows the amplitude-z parameter vs. A(z) function curve obtained at A0 = 30 nm for the PCL-PB-PCL and epoxy resin blend system. In Figure 6a, the solid line represents the values of the amplitude Ais(z) for an infinitely stiff sample assuming a zero indentation depth (δ). The A(z) curves observed for the PB and epoxy resin blend system lie above the Ais(z) curve because of the indentation on these samples. Since the deformation of the stiff silicon tip can be negligible compared with the sample indentation, the indentation depth δ(z) at a given z could be estimated from the equation δ(z) = A(z)-Ais(z) [13, 18]. The δ(rsp) curves were also obtained in a similar way. The δ(rsp) curves were plotted in Figure 7b. Finally, the phase shift-z distance curve and the phase shift-rsp curve were deduced from the amplitude-z and the phase angle-z curves, which are shown in Figures 6b and 6a, respectively.
It was found that the 'dark circle' shown in the height image and the phase image (Figure 5c, c') was close to the nanopores shown in the TEM image. So, the morphology and the phase separation observed in TM-AFM images could be verified.
The phase image contrast revealed in Figure 5 can be interpreted according to the phase shift-rsp curve. For rsp > 0.85, the phase shift is negative both on PB and matrix, and the phase shift difference between PB and matrix is small, which accounts for the obscure phase contrast obtained at rsp = 0.95 (shown in Figure 5a'). As rsp is in the region of 0.77 to 0.85, the phase shift becomes larger on the matrix than on PB, and the largest phase shift difference between PB and matrix was observed. So, in the phase images obtained at rsp = 0.77 (Figure 5c') and at rsp = 0.85 (Figure 5b'), the PB-rich region shows to be darker than the matrix region, and the phase images show maximum contrast. While the rsp decreases to 0.1 (Figure 5d'), a clear phase contrast is also obtained, and the PB-rich region remains to have a dark contrast. Additionally, it was found that the size of PB-rich domains decreases compared with that shown in Figure 5b', c' in both height and phase images. In the blend, the cross-linked epoxy resin of the matrix is a thermoset polymer, and the PB polymer of the domain materials is liquid at room temperature . The two materials are quite different in stiffness. So, it is necessary to take the indentation depth of the probe tip into consideration. The indentation depth-rsp curve (Figure 7b) reveals that the indentation depth of the PB-rich region is larger than that of the matrix when rsp is more than 0.8, and it is reverse when rsp is less than 0.8. For larger rsp, the tip-sample interaction on the matrix of PCL mixed with epoxy will be more affected by the attractive capillary force than on the PB domains because the matrix of epoxy resin is more hydrophilic than the PB domain due to the oxygen atoms. So, the stiffer matrix shows larger indentation depth. For smaller rsp, the indentation depth is nearly identical for the PB-rich domain and for the matrix, and the indentation depth is just about several nanometers. Considering that the radius size of the PB domain (approximately 10 nm) is close to the tip curvature radius (approximately 10 nm), when rsp is dramatically decreased, the tip contacts both the PB domain and the matrix simultaneously. The stiff edge of the domain will prevent the tip from indenting into the compliant PB-rich region. So, the size of the domain obtained is just the size of the tip, which results in the shrinkage of the measured size of the PB domain.
Many studies showed that to obtain images which describe the 'true' topography of a sample surface, the images should be recorded using sufficiently high set-point amplitudes and high rsp values. In this study, at rsp = 0.95 to 1.0, significant tip indentation in the PB region and the matrix did not take place (see Figure 7b). We thought that at this condition, the true topography is obtained (Figure 5a), but at rsp = 0.85, the height image contrast shown in Figure 5b was completely reverse. We found that in the phase shift-rsp curve, at rsp = 0.85, the interaction force between the tip and the sample varied from the attractive interaction on the PB into the repulsive interaction on the matrix (Figure 7a), leading to a large decrease in amplitude. Thus, the PB region looks higher than the real topography in the height image as far as the feedback mechanism is concerned. So, a height artifact is observed in Figure 5b, instead of the height image contrast inversion. While rsp ≤ 0.77, the tip-sample interaction force is repulsive for both the PB and the matrix, so the height images did not undergo reverse but influenced by the effect of tip indentation.
The above results demonstrated that it is important to select appropriate scanning parameters during TM-AFM measurement in order to get a good image with the clear morphology and topography information. The topography and nanoscale phase separation of PCL-PB-PCL-blended epoxy resin have been studied by tapping mode AFM at different rsp values and with different tips. The stiffer tips are found to be desired to get clear phase shift images. According to the force-probe measurements, the optimum condition for TM-AFM measurement is 0.77 ≤ rsp < 0.85. At lower rsp, the size of compliant PB-rich domains decreases because the tip size is close to the dimension of PB-rich domains. At rsp = 0.85, the tip-sample interaction force varies from the attractive interaction on the PB into the repulsive interaction on the matrix, which leads to the height image reverse phenomena.
For the reasonable interpretation of the TM-AFM phase and height images, it is necessary to take the indentation depth and the tip-sample interaction force into consideration. The present study is general and should be easily transferred for the analysis of a similar system.
- A 0 :
- A sp :
- r sp :
set-point amplitude ratio
transmission electron microscopy
tapping mode atomic force microscopy.
We would like to thank Professor Sixun Zheng's research group for the polymer samples. This work was partially supported by the National Natural Science Foundation of China (No. 10974134 and 31170961) and the National Key Basic Research Program (973 Project) (2010CB933900).
- Bar G, Thomann Y, Brandsch R, Cantow HJ, Whangbo MH: Factors affecting the height and phase images in tapping mode atomic force microscopy. Study of phase-separated polymer blends of poly(ethene-co-styrene) and poly(2,6-dimethyl-1,4-phenylene oxide). Langmuir 1997, 13: 3807–3812. 10.1021/la970091mView ArticleGoogle Scholar
- Bar G, Thomann Y, Whangbo MH: Characterization of the morphologies and nanostructures of blends of poly(styrene) block-poly(ethene-co-but-1-ene)-block-poly(styrene) with isotactic and atactic polypropylenes by tapping-mode atomic force microscopy. Langmuir 1998, 14: 1219–1226. 10.1021/la9711544View ArticleGoogle Scholar
- McLean RS, Sauer BB: Tapping-mode AFM studies using phase detection for resolution of nanophases in segmented polyurethanes and other block copolymers. Macromolecules 1997, 30: 8314–8317. 10.1021/ma970350eView ArticleGoogle Scholar
- Wang Y, Song R, Li YS, Shen JS: Understanding tapping-mode atomic force microscopy data on the surface of soft block copolymers. Surf Sci 2003, 530: 136–148. 10.1016/S0039-6028(03)00388-1View ArticleGoogle Scholar
- Garcia R, Perez R: Dynamic atomic force microscopy methods. Surf Sci Rep 2002, 47: 197–301. 10.1016/S0167-5729(02)00077-8View ArticleGoogle Scholar
- Godehardt R, Lebek W, Adhikari R, Rosenthal M, Martin C, Frangov S, Michler G: Optimum topographical and morphological information in AFM tapping mode investigation of multicomponent polyethylene. Eur Polymer J 2004, 40: 917–926. 10.1016/j.eurpolymj.2004.01.034View ArticleGoogle Scholar
- Lin ZQ, Kim DH, Wu XD, Boosahda L, Stone D, LaRose L, Russell TP: A rapid route to arrays of nanostructures in thin films. Adv Mater 2002, 14: 1373–1376. 10.1002/1521-4095(20021002)14:19<1373::AID-ADMA1373>3.0.CO;2-FView ArticleGoogle Scholar
- Sanchez MS, Mateo JM, Colomer FJR, Ribelles JLG: Nanoindentation and tapping mode AFM study of phase separation in poly(ethyl acrylate-co-hydroxyethyl methacrylate) copolymer networks. Eur Polymer J 2006, 42: 1378–1383. 10.1016/j.eurpolymj.2005.12.018View ArticleGoogle Scholar
- Ye Q, Wang Y, Spencer P: Nanophase separation of polymers exposed to simulated bonding conditions. J Biomed Mater Res Part B-Appl Biomater 2009, 88B: 339–348. 10.1002/jbm.b.31047View ArticleGoogle Scholar
- Ogoshi T, Fujiwara T, Bertolucci M, Galli G, Chiellini E, Chujo Y, Wynne KJ: Tapping mode AFM evidence for an amorphous reticular phase in a condensation-cured hybrid elastomer: alpha, omega-dihydroxypoly(dimethylsiloxane)/poly(diethonysiloxane)/fumed silica nanoparticles. J Am Chem Soc 2004, 126: 12284–12285. 10.1021/ja0474345View ArticleGoogle Scholar
- Paredes JI, Martinez-Alonso A, Tascon JMD: Detecting surface oxygen groups on carbon nanofibers by phase contrast imaging in tapping mode AFM. Langmuir 2003, 19: 7665–7668. 10.1021/la020880qView ArticleGoogle Scholar
- Ruozi B, Tosi G, Tonelli M, Bondioli L, Mucci A, Forni F, Vandelli MA: AFM phase imaging of soft-hydrated samples: a versatile tool to complete the chemical-physical study of liposomes. J Liposome Res 2009, 19: 59–67. 10.1080/08982100802584071View ArticleGoogle Scholar
- Bar G, Ganter M, Brandsch R, Delineau L, Whangbo MH: Examination of butadiene/styrene-co-butadiene rubber blends by tapping mode atomic force microscopy. Importance of the indentation depth and reduced tip-sample energy dissipation in tapping mode atomic force microscopy study of elastomers. Langmuir 2000, 16: 5702–11. 10.1021/la9913699View ArticleGoogle Scholar
- Bar G, Brandsch R, Whangbo MH: Effect of tip sharpness on the relative contributions of attractive and repulsive forces in the phase imaging of tapping mode atomic force microscopy. Surf Sci 1999, 422: L192-L9. 10.1016/S0039-6028(98)00899-1View ArticleGoogle Scholar
- Brandsch R, Bar G, Whangbo MH: On the factors affecting the contrast of height and phase images in tapping mode atomic force microscopy. Langmuir 1997, 13: 6349–6353. 10.1021/la970822iView ArticleGoogle Scholar
- Chen X, Roberts CJ, Zhang J, Davies MC, Tendler SJB: Phase contrast and attraction-repulsion transition in tapping mode atomic force microscopy. Surf Sci 2002, 519: L593-L598. 10.1016/S0039-6028(02)02217-3View ArticleGoogle Scholar
- Chen X, Davies MC, Roberts CJ, Tendler SJB, Williams PM, Burnham NA: Optimizing phase imaging via dynamic force curves. Surf Sci 2000, 460: 292–300. 10.1016/S0039-6028(00)00574-4View ArticleGoogle Scholar
- Bar G, Delineau L, Brandsch R, Bruch M, Whangbo MH: Importance of the indentation depth in tapping-mode atomic force microscopy study of compliant materials. Appl Phys Lett 1999, 75: 4198–4200. 10.1063/1.125581View ArticleGoogle Scholar
- Raghavan D, Gu X, Nguyen T, VanLandingham M, Karim A: Mapping polymer heterogeneity using atomic force microscopy phase imaging and nanoscale indentation. Macromolecules 2000, 33: 2573–2583. 10.1021/ma991206rView ArticleGoogle Scholar
- Jiyun F, Lu-Tao W, Chi-Ming C, Jie X, Lin L: Imaging of sub-surface nano particles by tapping-mode atomic force microscopy. Polymer 2001, 42: 2259–2262. 10.1016/S0032-3861(00)00464-XView ArticleGoogle Scholar
- Spitzner EC, Riesch C, Magerle R: Subsurface imaging of soft polymeric materials with nanoscale resolution. Acs Nano 2011, 5: 315–320. 10.1021/nn1027278View ArticleGoogle Scholar
- Meng FL, Zheng SX, Zhang W, Li HQ, Liang Q: Nanostructured thermosetting blends of epoxy resin and amphiphilic poly(epsilon-caprolactone)-block-polybutadiene-block-poly(epsilon-caprolactone) triblock copolymer. Macromolecules 2006, 39: 711–719. 10.1021/ma0518499View ArticleGoogle Scholar
- Huckstadt H, Gopfert A, Abetz V: Synthesis and morphology of ABC heteroarm star terpolymers of polystyrene, polybutadiene and poly(2-vinylpyridine). Macromol Chem Phys 2000, 201: 296–307. 10.1002/(SICI)1521-3935(20000201)201:3<296::AID-MACP296>3.0.CO;2-UView ArticleGoogle Scholar
- Thormann E, Pettersson T, Kettle J, Claesson PM: Probing material properties of polymeric surface layers with tapping mode AFM: which cantilever spring constant, tapping amplitude and amplitude set point gives good image contrast and minimal surface damage? Ultramicroscopy 2010, 110: 313–319. 10.1016/j.ultramic.2010.01.002View ArticleGoogle Scholar
- Bhushan B, Qi J: Phase contrast imaging of nanocomposites and molecularly thick lubricant films in magnetic media. Nanotechnology 2003, 14: 886–895. 10.1088/0957-4484/14/8/309View ArticleGoogle Scholar
- Han X, Xu J, Liu HL, Hu Y: Image contrast inversion of a solvent cast SEBS film. Chin J Chem 2006, 24: 149–52. 10.1002/cjoc.200690013View ArticleGoogle Scholar
- Wang H, Djurisic AB, Chan WK, Xie MH: Factors affecting phase and height contrast of diblock copolymer PS-b-PEO thin films in dynamic force mode atomic force microscopy. Appl Surf Sci 2005, 252: 1092–1100. 10.1016/j.apsusc.2005.02.034View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.