The use of artificial neural networks in electrostatic force microscopy
© Castellano-Hernandez et al.; licensee Springer 2012
Received: 6 January 2012
Accepted: 15 May 2012
Published: 15 May 2012
The use of electrostatic force microscopy (EFM) to characterize and manipulate surfaces at the nanoscale usually faces the problem of dealing with systems where several parameters are not known. Artificial neural networks (ANNs) have demonstrated to be a very useful tool to tackle this type of problems. Here, we show that the use of ANNs allows us to quantitatively estimate magnitudes such as the dielectric constant of thin films. To improve thin film dielectric constant estimations in EFM, we first increase the accuracy of numerical simulations by replacing the standard minimization technique by a method based on ANN learning algorithms. Second, we use the improved numerical results to build a complete training set for a new ANN. The results obtained by the ANN suggest that accurate values for the thin film dielectric constant can only be estimated if the thin film thickness and sample dielectric constant are known.
PACS: 07.79.Lh; 07.05.Mh; 61.46.Fg.
KeywordsElectrostatic force microscopy Thin films Artificial neural networks
When electrostatic force microscopy (EFM) [1–6] is working at the nanoscale, several interacting parameters have a strong influence in the signal . Since the electrostatic force is a long-range interaction, macroscopic parameters such as the shape of the tip or the sample thickness can strongly modify the electrostatic interaction [8, 9]. However, in many experimental situations, it is not possible to obtain accurate values for all of these parameters, and it is very difficult to achieve quantitative experimental results . Previous results  have shown that artificial neural networks (ANNs)  are a useful tool to characterize dielectric samples in highly undetermined EFM systems. Using known force vs. distance curves as inputs for their training, ANNs have been able to estimate the dielectric constant of a semi-infinite sample in a system where the tip radius and shape were not known.
In this paper, we demonstrate that ANNs can be used to improve the accuracy of numerical simulations in EFM and to quantitatively estimate the thin film dielectric constant from vertical force curves. First, we compare standard minimization and ANN techniques, demonstrating that ANN techniques provide a better control of the final result of the simulation. The improved numerical results are also used to create a complete training set of an ANN that estimates the dielectric constant of a thin film placed over a dielectric sample.
As it has been shown before , ANNs are able to estimate physical magnitudes in highly undetermined systems. In this article, we train an ANN with a complete thin film sample to study the necessity of knowing the geometry of the sample in the estimations of the thin film dielectric constant. Although the influence of the thin film thickness is much larger than that of the substrate dielectric constant, we demonstrate that accurate values of the thin film dielectric constant can only be obtained when both magnitudes are known.
Artificial neural network formalism for the calculation of electric fields
Coefficients obtained by the ANN and LSM algorithms for an EFM system
Results and discussion
Thin film dielectric constant estimation
The ANN can be used with realistic experimental curves without any previous treatment, which is one of the advantages of using this technique . In this case, experimental curves with a high error could make the ANN give wrong ϵ1 estimations. This problem can be easily solved by training the ANN with a mixture of experimental and numerical F vs. D curves. This strategy would make the ANN more robust against experimental noise (by the use of experimental curves) and still effective on the ϵ1 estimations (by the use of a whole set of numerical curves).
Recently, a simple analytical expression has been developed that demonstrates that a sample composed by a thin film over a dielectric substrate gives the same response as that of a semi-infinite uniform dielectric sample . The fact that different combinations of ϵ1ϵ2, and h1 can correspond to the same effective dielectric constant is in agreement with the results found in Figure 4a since including ϵ2 and h1 as input values improves the ANN performance in the ϵ1 estimations.
We have demonstrated that ANNs can strongly improve the efficiency of the characterization of samples by electrostatic force microscopy. First, we have demonstrated that the generalized image charge method can be modified to use a neural network minimization algorithm. Using this technique, we have increased the accuracy of the electrostatic force and capacitance calculations. By using electrostatic force simulations, we have been able to train an ANN to estimate the dielectric constant of thin films. The analysis of the results of the ANN suggests that the thin film dielectric constant can only be obtained when the thin film thickness and the dielectric nature of the sample are known. Note that the methods explained in this paper can be easily applied to experimental data by providing this kind of input to the ANN. If enough data are available, experimental curves can be used for the ANN training alone or together with theoretical curves.
Artificial neural networks
Electrostatic force microscopy
- F vs. D:
Force vs. tip-sample distance
Generalized image charge method
This work was supported by TIN2010-19607 and BFU2009-08473. GMS acknowledges support from the Spanish Ramón y Cajal Program.
- Kalinin SV, Jesse S, Rodriguez BJ, Eliseev EA, Gopalan V, Morozovska AN: Quantitative determination of tip parameters in piezoresponse force microscopy. Appl Phys Lett 2007, 90: 212905. 10.1063/1.2742900View ArticleGoogle Scholar
- Lyuksyutov SF, Vaia RA, Paramonov PB, Juhl S, Waterhouse L, Ralich RM, Sigalov G, Sancaktar E: Electrostatic nanolithography in polymers using atomic force microscopy. Nat Mater 2003, 2: 468–472. 10.1038/nmat926View ArticleGoogle Scholar
- Guriyanova S, Golovko DS, Bonaccurso E: Cantilever contribution to the total electrostatic force measured with the atomic force microscope. Measurement Science & Technology 2010, 21: 025502. 10.1088/0957-0233/21/2/025502View ArticleGoogle Scholar
- Palacios-Lidon E, Abellan J, Colchero J, Munuera C, Ocal C: Quantitative electrostatic force microscopy on heterogeneous nanoscale samples. Appl Phys Lett 2005, 87: 154106. 10.1063/1.2099527View ArticleGoogle Scholar
- Hu J, Xiao XD, Salmeron M: Scanning polarization force microscopy - a technique for imaging liquids and weakly adsorbed layers. Appl Phys Lett 1995, 67: 476–478. 10.1063/1.114541View ArticleGoogle Scholar
- Morozovska AN, Eliseev EA, Kalinin SV: The piezoresponse force microscopy of surface layers and thin films: effective response and resolution function. J Appl Phys 2007, 102: 074105. 10.1063/1.2785824View ArticleGoogle Scholar
- Butt HJ, Cappella B, Kappl M: Force measurements with the atomic force microscope: technique, interpretation and applications. Surf Sci Rep 2005, 59: 1–152. 10.1016/j.surfrep.2005.08.003View ArticleGoogle Scholar
- Sacha GM, Gomez-Navarro C, Saenz JJ, Gomez-Herrero J: Quantitative theory for the imaging of conducting objects in electrostatic force microscopy. Appl Phys Lett 2006, 89: 173122. 10.1063/1.2364862View ArticleGoogle Scholar
- Sacha GM, Saenz JJ: Cantilever effects on electrostatic force gradient microscopy. Appl Phys Lett 2004, 85: 2610–2612. 10.1063/1.1797539View ArticleGoogle Scholar
- Sacha GM, Verdaguer A, Martinez J, Saenz JJ, Ogletree DF, Salmeron M: Effective tip radius in electrostatic force microscopy. Appl Phys Lett 2005, 86: 123101. 10.1063/1.1884764View ArticleGoogle Scholar
- Sacha GM, Rodriguez FB, Varona P: An inverse problem solution for undetermined electrostatic force microscopy setups using neural networks. Nanotechnology 2009, 20: 085702. 10.1088/0957-4484/20/8/085702View ArticleGoogle Scholar
- Haykin S: Feedforward Neural Networks: An Introduction. Prentice-Hall, Englewood; 1999.Google Scholar
- Hänninen JJ, Lindell IV, Nikoskinen KI: Electrostatic image theory for an anisotropic boundary of an anisotropic half-space. Progress in Electromagnetics Research-Pier 2004, 47: 236–262.View ArticleGoogle Scholar
- Sacha GM, Sahagun E, Saenz JJ: A method for calculating capacitances and electrostatic forces in atomic force microscopy. J Appl Phys 2007, 101: 024310. 10.1063/1.2424524View ArticleGoogle Scholar
- Sacha GM: Página de Sacha. [www.ii.uam.es/~sacha] 
- Tetko IV, Livingstone DJ, Luik AI: Neural-network studies.1.Comparison of overfitting and overtraining. Journal of Chemical Information and Computer Sciences 1995, 35: 826–833. 10.1021/ci00027a006Google Scholar
- Sacha GM, Rodriguez FB, Serrano E, Varona P: Generalized image charge method to calculate electrostatic magnitudes at the nanoscale powered by artificial neural networks. Journal of Electromagnetic Waves and Applications 2010, 24: 1145–1155. 10.1163/156939310791586160View ArticleGoogle Scholar
- Castellano-Hernández E, Sacha GM: Ultrahigh dielectric constant of thin films obtained by electrostatic force microscopy and artificial neural networks. Applied Physics Letters 2012, 100: 023101. 10.1063/1.3675446View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.