- Nano Ideas
- Open Access
A Scheme for Solving the Plane–Plane Challenge in Force Measurements at the Nanoscale
© The Author(s) 2010
- Received: 23 March 2010
- Accepted: 4 May 2010
- Published: 19 May 2010
Non-contact interaction between two parallel flat surfaces is a central paradigm in sciences. This situation is the starting point for a wealth of different models: the capacitor description in electrostatics, hydrodynamic flow, thermal exchange, the Casimir force, direct contact study, third body confinement such as liquids or films of soft condensed matter. The control of parallelism is so demanding that no versatile single force machine in this geometry has been proposed so far. Using a combination of nanopositioning based on inertial motors, of microcrystal shaping with a focused-ion beam (FIB) and of accurate in situ and real-time control of surface parallelism with X-ray diffraction, we propose here a “gedanken” surface-force machine that should enable one to measure interactions between movable surfaces separated by gaps in the micrometer and nanometer ranges.
- Nanoscale interactions
- Plane–plane geometry
- Surface-force machine
- X-ray diffraction
Measurements of non-contact interactions between surfaces have always been a challenge. This includes the presence of a third body (e.g. liquid or gas) in the separating gap, with gap varying from micrometers down to the nanoscale. In the context of complex fluids, precise measurements between extended and curved surfaces have been done using the Surface Force Apparatus (SFA) [1–3]. In any environment, such as vacuum, gas and liquid, the measurement of interactions between nanoobjects has greatly benefited from the Atomic Force Microscope (AFM) [4–11]. Control and resolution in AFM interaction measurement has reached picoNewton scale in surface imaging under ultra-high vacuum. Cryogenic environment for single electron-spin detection even pushed the limit down to the attoNewton scale .
The use of the plane–plane geometry for measuring nanoscale interactions remains somewhat as a dream as this is the simplest geometry used in many models. Indeed, this geometry enables exact calculations. In addition, flat surfaces are easier to control at the nanoscale (roughness, contamination, chemical functionalization, surface patterning). Measurements in the plane–plane geometry have been attempted in the framework of Casimir force studies, in order to probe the theoretical predictions of mechanical effects related to quantum vacuum fluctuations [2, 3, 13]. The limited accuracy obtained in historical Casimir force measurements based on macroscopic instrumental controls stresses the extraordinary difficulty of this sort of measurements (the reported agreement between theory and experimental data in  is ≈15%). This is mostly due to difficulties in controlling the parallelism of flat surfaces separated by gaps falling in the micro- or nano-range with enough accuracy.
In contrast, measurements in the commonly used sphere-plane geometry [4–11] are in principle easier since a flat surface and a facing sphere do not need to be made parallel. However, interpreting such measurements immediately implies the use of the Derjaguin approximation, often referred to as the Proximity Force Approximation (PFA), whose validity has always been a matter of endless debates .
Previous attempts of plane–plane interaction force instruments were closely related to the SFA [2, 3]. The misalignment between the planes surfaces was controlled using capacitive forces between two rotating metallic plates related to the samples. The precision allowed by this kind of set-up on the angle control was ≈2 × 10−3 deg . An unprecedented precision in the parallelism between plane surfaces has been achieved by the Nesvizhevsky’s group at the Institut Laue Langevin . During the measurement of the quantum states of neutrons in the Earth gravitational field, these authors were able to align two macroscopic planes (10 cm size) with an angular precision of ≈10−4 deg. However none force measurement set-up can be implemented in their experiment.
In this paper we analyze how the combined use of (i) inertial motors for nano-positioning (translation and rotation), (ii) nano-tools such as FIB and (iii) X-ray diffraction on single crystal for real time, in situ alignment control can overcome the key difficulties in the design of a plane–plane Surface Force Machine (p2SFM). We shall see that: (i) inertial motors originally designed for low-temperature scanning-probe microscopy (SPM) (see e.g. [16, 17]) are used both to control and vary the distance d between interacting surfaces and their relative orientation (θ, φ); (ii) FIB is used to precisely weld a flat silicon single crystal at the extremity of an AFM cantilever and (iii) the sharpness of X-ray diffraction at Bragg position is sufficient to control parallelism and that it can be implemented so that this control takes place in situ and in real time.
When surfaces are kept at micron or sub-micron distances from each other, interaction phenomena, generally neglected at the macroscopic scale, take place between them. At the submicron scale major interaction forces between surfaces are:
· Electrostatic forces;
· Hydrodynamic forces mediated by the confined fluid environment;
· Near-field radiative heat exchanges;
· Van der Waals and Casimir forces.
If we consider the plane–plane configuration introduced above, it is possible to define the dependence of the various interactions on the distance d between surfaces. Choosing to list the interactions from the weakest to the strongest dependence on distance, we have:
with A the interacting surface, η the fluid viscosity and v the relative velocity between plates;
· Electrostatic force between two conductors:
with V the voltage drop between the conductors, V O the residual potential between plates, A their interacting surface, and ε = ε r ε 0 the medium permittivity;
with w and L the dimensions of the plate, η the fluid viscosity, and v the relative velocity between plates;
· Casimir force between two perfect mirrors :
with the Plank’s constant, c the speed of light in vacuum, and A the plate surface.
In a recent paper  we presented a comparison between experiments and theory for the hydrodynamic force with perfect slip boundary conditions  between a flat AFM cantilever and a plane substrate. The cantilever oscillated in air while the plane substrate was approached from hundreds microns down to hundreds nanometers. The plane substrate was mounted over three linear inertial motors based on stick-and-slip technology allowing a millimeter range displacement (8 mm) with nanometer scale resolution (≈40 nm/step). The agreement between experiment and theory obtained solving Navier-Stokes equation together with perfect slip boundary conditions is 5%, when a residual misalignment of 0.7 deg is considered. The experimental set-up developed for the measurement did not offer any misalignment correction. However the residual misalignment between the two planes does not preclude a comparison between theory and experiment. This is because of the weak distance dependence of the interaction force in analysis.
During the positioning of the block, the precision in the angle that can be achieved is in the order of 0.1 deg. Furthermore, the block surface can also be polished using FIB to obtain a better quality of the surface (roughness less than 10 nm r.m.s.). It is known that a FIB treatment can induce surface charging. Although it should not be a problem for hydrodynamics, thermal transfer and even electrostatic measurements, this is surely unacceptable in a quantitative Casimir force measurement. In this case, gold metalization of the crystal surface will certainly be needed.
The positioning obtained after the insertion of the two surfaces and the force measurement system is not better than 0.1 deg. This is not enough for interaction measurements in the plane–plane geometry. Beside changing and controlling the gap size, it is also necessary to change and control the relative orientation of the two surfaces. For this purpose, the two interacting surfaces are put on different mechanical stages that have to be approached each other. The approach can be performed using a linear translator based on inertial motors. To achieve the required precision in parallelism, a two-angle tilter system has to be implemented in the set-up. Micro-goniometers, that have been recently developed, allow one to adjust angle with a precision, at room temperature, better than 10−4 deg (see for example ).
Considering the scheme in figure 3 we note that the second surface satisfies the Bragg condition only when it is perfectly parallel to the first surface (see below). This technique allows then to control the parallelism between surfaces by recording the evolution of the out-coming X-ray beam intensity as a function of the relative orientation (θ, φ) between the interacting surfaces. The out coming beam intensity reaches its maximum value when the lattice vectors of the two surfaces are parallel. Using a single information it is then possible to control both angles (θ, φ). The precision that can be achieved using such a control procedure is given by the Rocking curve of the selected materials.
When the separation between the two surfaces enters in the micron and submicron range multiple reflections can undergo inside the cavity, thereby affecting the intensity of the outgoing X-ray beam. Considering that approximately 5% of the intensity is lost per reflection one should limit the multiple reflections within the order of 10 to maintain S/N in excess of 50 and not deteriorating the precision in parallelism. This gives a limit in the smallest attainable gap that depends on the surface size and the incidence angle. In the case of Si (3 3 3), with a surface size of 50 × 50 μm2, the minimum gap is in the order of 200 nm. Such value is the typical gap size attained for Casimir force measurements.
It is known that a mismatch between the surface and the lattice axes can exist. The proposed procedure guarantees a very good alignment between the lattice axes that cannot be directly translated to a good surface alignment. Coupling optical reflection of a laser beam with diffraction of an X-ray beam will allow determining very precisely the orientation of the surface with respect to the lattice axes. Furthermore, and most important, if the two samples are cut from the same wafer the mismatch will be the same for both surfaces. Then, the good alignment of the lattice will be translated to a good surface alignment.
It is worth noticing additionally that the proposed alignment procedure is very general and not limited to the alignment of two surfaces made of a single material. It is possible to cover a Silicon single crystal with a thin film (t < 1 μm) of a different material. A thin film is basically transparent to X-ray beams  so that diffraction occurs at the level of the Silicon single crystals. It is then clear that depositing films of different materials is possible without affecting the validity of the procedure proposed here.
The precision in the separation and parallelism between two surfaces that can be obtained by the here proposed p2SFM will allow one to make a reliable measurement of the interactions between two plane surfaces. The precision in parallelism, in particular, is sufficient to measure also the interaction force that exhibits the strongest dependence on the distance, i.e., the Casimir force. As pointed out in the introduction, the use of the plane–plane geometry will allow avoiding using the Derjaguin approximation (PFA) made for the sphere-plane configuration . Using the here proposed p2SFM, it is even possible, in principle, to test the limit of validity of the PFA. Starting from the case of perfect parallel planes one can increase the misalignment angle and compare the experimental results with the theoretical model obtain in the PFA framework. It should then be possible to find a critical angle setting the limit of validity.
In summary, we have first recalled the performances in orientation control that are required to perform interaction force measurements in the plane–plane geometry at the nanoscale. In order to reliably design a plane–plane surface force machine, we have then proposed an original combination of existing elements originating from different fields of instrumentation. The proposed force machine is sketched in figure 5.
Beside classical temperature and distance controls, we shall use:
• A combination of oriented flat surfaces and a lever for force measurements that is based on FIB, a key tool in nanotechnology (see fig. 2);
• Nanopositioning that has been originally developed for scanning-probe microscopy and that is based on inertial motors;
• A precise control of orientation based on X-ray diffraction on high quality single crystals that is routinely used at the required precision in synchrotron facilities.
Although the set-up proposed here has not been realized and tested up to now, we believe that its design will give birth to a new generation of versatile and original force machines dedicated to the investigation on non-contact interactions between surfaces in the plane–plane geometry and at the nanoscale.
This article is distributed under the terms of the Creative Commons Attribution Noncommercial License which permits any noncommercial use, distribution, and reproduction in any medium, provided the original author(s) and source are credited.
- Cottin-Bizonne C, Cross B, Steinberger A, Charlaix E: Boundary slip on smooth hydrophobic surfaces: intrinsic effects and possible artifacts. Phys. Rev. Lett. 2004, 94: 056102. Bibcode number [2005PhRvL..94e6102C] Bibcode number [2005PhRvL..94e6102C] 10.1103/PhysRevLett.94.056102View ArticleGoogle Scholar
- Lamoreaux SK: Demonstration of the Casimir force in the 0.6 to 6 μm range, Phys. Rev. Lett. 1997, 78: 5–8. COI number [1:CAS:528:DyaK2sXjvV2rtA%3D%3D]; Bibcode number [1997PhRvL..78....5L] COI number [1:CAS:528:DyaK2sXjvV2rtA%3D%3D]; Bibcode number [1997PhRvL..78....5L] 10.1103/PhysRevLett.78.5View ArticleGoogle Scholar
- Bressi G, Carugno G, Onofrio R, Ruoso G: Measurement of the Casimir force between parallel metallic surfaces. Phys. Rev. Lett. 2002, 88: 041804. COI number [1:STN:280:DC%2BD387hs1ajtQ%3D%3D]; Bibcode number [2002PhRvL..88d1804B] COI number [1:STN:280:DC%2BD387hs1ajtQ%3D%3D]; Bibcode number [2002PhRvL..88d1804B] 10.1103/PhysRevLett.88.041804View ArticleGoogle Scholar
- Mohideen U, Roy A: Precision measurement of the Casimir force from 0.1 to 0.9 μm. Phys. Rev. Lett. 1998, 81: 4549–4552. COI number [1:CAS:528:DyaK1cXnsVOjtLw%3D]; Bibcode number [1998PhRvL..81.4549M] COI number [1:CAS:528:DyaK1cXnsVOjtLw%3D]; Bibcode number [1998PhRvL..81.4549M] 10.1103/PhysRevLett.81.4549View ArticleGoogle Scholar
- Jourdan G, Lambrecht A, Comin F, Chevrier J: Quantitative non-contact dynamic Casimir force measurements. EPL 2009, 85: 31001. Bibcode number [2009EL.....8531001J] Bibcode number [2009EL.....8531001J] 10.1209/0295-5075/85/31001View ArticleGoogle Scholar
- Munday JN, Capasso F: Precision measurement of the Casimir-Lifshitz force in a fluid. Phys. Rev. A 2007, 75: 060102. Bibcode number [2007PhRvA..75f0102M] Bibcode number [2007PhRvA..75f0102M] 10.1103/PhysRevA.75.060102View ArticleGoogle Scholar
- Munday JN, Capasso F, Parsegian VA, Bezrukov SM: Measurements of the Casimir-Lifshitz force in fluids: The effect of electrostatic forces and Debye screening. Phys. Rev. A 2008, 78: 032109. Bibcode number [2008PhRvA..78c2109M] Bibcode number [2008PhRvA..78c2109M] 10.1103/PhysRevA.78.032109View ArticleGoogle Scholar
- Munday JN, Capasso F, Parsegian VA: Measured long-range repulsive Casimir–Lifshitz forces. Nature 2009, 457: 170–173. COI number [1:CAS:528:DC%2BD1MXitlSiuw%3D%3D]; Bibcode number [2009Natur.457..170M] COI number [1:CAS:528:DC%2BD1MXitlSiuw%3D%3D]; Bibcode number [2009Natur.457..170M] 10.1038/nature07610View ArticleGoogle Scholar
- De Man S, Heeck K, Wijngaarden RJ, Iannuzzi D: Halving the Casimir force with conductive oxides. Phys Rev. Lett. 2009, 103: 040402. 10.1103/PhysRevLett.103.040402View ArticleGoogle Scholar
- De Man S, Heeck K, Iannuzzi D: No anomalous scaling in electrostatic calibrations for Casimir force measurements. Phys. Rev. A 2009, 79: 024102. Bibcode number [2009PhRvA..79b4102D] Bibcode number [2009PhRvA..79b4102D] 10.1103/PhysRevA.79.024102View ArticleGoogle Scholar
- Rousseau E, Siria A, Jourdan G, Voltz S, Comin F, Chevrier J, Greffet JJ: Radiative heat transfer at the nanoscale. Nat. Photon. 2009, 3: 514–517. COI number [1:CAS:528:DC%2BD1MXhtVGgs7nE]; Bibcode number [2009NaPho...3..514R] COI number [1:CAS:528:DC%2BD1MXhtVGgs7nE]; Bibcode number [2009NaPho...3..514R] 10.1038/nphoton.2009.144View ArticleGoogle Scholar
- Rugar D, Budakian R, Mamin HJ, Chui BW: Single spin detection by magnetic resonance force microscopy. Nature 2004, 430: 329–332. COI number [1:CAS:528:DC%2BD2cXls1eqs78%3D]; Bibcode number [2004Natur.430..329R] COI number [1:CAS:528:DC%2BD2cXls1eqs78%3D]; Bibcode number [2004Natur.430..329R] 10.1038/nature02658View ArticleGoogle Scholar
- Casimir HBG: On the attraction between two perfectly conducting plates. Proc. Kon. Akad. Wet. 1948, 51: 793–795.Google Scholar
- Derjaguin BV, Abrikosova II, Lifshitz EM: Direct measurement of molecular attraction between solids separated by a narrow gap. Q. Rev. Chem. Soc. 1956, 10: 295–329. 10.1039/qr9561000295View ArticleGoogle Scholar
- Nesvizhevsky VV, Börner HG, Petukhov AK, Abele H, Bässler S, Rüss FJ, Störferle T, Westphal A, Gagarski AM, Petrov GA, Strelkov AV: Quantum states of neutrons in the Earth’s gravitational field. Nature 2002, 415: 297–299. Bibcode number [2002Natur.415..297N] Bibcode number [2002Natur.415..297N] 10.1038/415297aView ArticleGoogle Scholar
- Obermüller C, Deisenrieder A, Abstreiter G, Karrai K, Grosse S, Manus S, Feldmann J, Lipsanen H, Sopanen M, Ahopelto J: Pauli-blocking imaging of single strain-induced semiconductor quantum dots. Appl. Phys. Lett. 1999, 74: 3200–3202. Bibcode number [1999ApPhL..74.3200O] Bibcode number [1999ApPhL..74.3200O] 10.1063/1.124116View ArticleGoogle Scholar
- Brun M, Huant S, Woehl JC, Motte JF, Marsal L, Mariette H: Low-temperature near-field spectroscopy of CdTe quantum dots. J. Microscopy 2001, 202: 202–208. COI number [1:CAS:528:DC%2BD3MXjslSqtrs%3D] COI number [1:CAS:528:DC%2BD3MXjslSqtrs%3D] 10.1046/j.1365-2818.2001.00798.xView ArticleGoogle Scholar
- Siria A, Drezet A, Marchi F, Comin F, Huant S, Chevrier J: Viscous cavity damping of a microlever in a simple fluid. Phys. Rev. Lett 2009, 102: 254503. COI number [1:STN:280:DC%2BD1MrjvFGqtw%3D%3D]; Bibcode number [2009PhRvL.102y4503S] COI number [1:STN:280:DC%2BD1MrjvFGqtw%3D%3D]; Bibcode number [2009PhRvL.102y4503S] 10.1103/PhysRevLett.102.254503View ArticleGoogle Scholar
- Tabeling P: Introduction to microfluidics. Oxford University Press, New York; 2006.Google Scholar
- Drezet A, Siria A, Huant S, Chevrier J: Giant slip lengths of a simple fluid at vibrating solid interfaces. Phys. Rev. E 2010, 81: 046315. Bibcode number [2010PhRvE..81d6315D] Bibcode number [2010PhRvE..81d6315D] 10.1103/PhysRevE.81.046315View ArticleGoogle Scholar
- Rousseau E, Laroche M, Greffet JJ: Radiative heat transfer at nanoscale mediated by surface plasmons for highly doped silicon. Appl. Phys. Lett.. 2009, 95: 231913.Google Scholar
- Mulet JP: Modlisation du rayonnement thermique par une approche lectromagnetique. Rle des ondes de surface dans le tranfert d’nergie aux courtes chelles et dans les forces de Casimir. In Ph.D Dissertation. Universit Paris Sud – Paris XI, Paris; 2003.Google Scholar
- Joulain K, Mulet JP, Marquier F, Carminati R, Greffet JJ: Surface electromagnetic waves thermally excited: Radiative heat transfer, coherence properties and Casimir forces revisited in the near field. Surf. Sci. Rep. 2005, 57: 59–112. COI number [1:CAS:528:DC%2BD2MXjs1CmtLY%3D]; Bibcode number [2005SurSR..57...59J] COI number [1:CAS:528:DC%2BD2MXjs1CmtLY%3D]; Bibcode number [2005SurSR..57...59J] 10.1016/j.surfrep.2004.12.002View ArticleGoogle Scholar
- Mulet JP, Joulain K, Carminati R, Greffet JJ: Enhanced radiative heat transfer at nanometric distances. Micro. Thermophys. Eng. 2002, 6: 209–222. 10.1080/10893950290053321View ArticleGoogle Scholar
- Volokitin AI, Persson BN: Near-field radiative heat transfer and noncontact friction. Rev. Mod. Phys. 2007, 79: 1291–1329. COI number [1:CAS:528:DC%2BD1cXisFCjtL8%3D]; Bibcode number [2007RvMP...79.1291V] COI number [1:CAS:528:DC%2BD1cXisFCjtL8%3D]; Bibcode number [2007RvMP...79.1291V] 10.1103/RevModPhys.79.1291View ArticleGoogle Scholar
- Mulet JP, Joulain K, Carminati R, Greffet JJ: Nanoscale radiative heat transfer between a small particle and a plane surface. Appl. Phys. Lett. 2000, 78: 2931–2933. Bibcode number [2001ApPhL..78.2931M] Bibcode number [2001ApPhL..78.2931M] 10.1063/1.1370118View ArticleGoogle Scholar
- Henkel C, Joulain K, Carminati R, Greffet JJ: Spatial coherence of thermal near fields. Opt. Commun. 2000, 186: 57–67. COI number [1:CAS:528:DC%2BD3cXosFGht7w%3D]; Bibcode number [2000OptCo.186...57H] COI number [1:CAS:528:DC%2BD3cXosFGht7w%3D]; Bibcode number [2000OptCo.186...57H] 10.1016/S0030-4018(00)01048-8View ArticleGoogle Scholar
- Zachariasen WH: Theory of X-Ray diffraction in crystals. Dover Publications Inc, Mineola; 2004.Google Scholar