- Nano Express
- Open Access
2D array of cold-electron nanobolometers with double polarised cross-dipole antennas
© Kuzmin; licensee Springer. 2012
- Received: 1 November 2011
- Accepted: 3 March 2012
- Published: 18 April 2012
A novel concept of the two-dimensional (2D) array of cold-electron nanobolometers (CEB) with double polarised cross-dipole antennas is proposed for ultrasensitive multimode measurements. This concept provides a unique opportunity to simultaneously measure both components of an RF signal and to avoid complicated combinations of two schemes for each polarisation. The optimal concept of the CEB includes a superconductor-insulator-normal tunnel junction and an SN Andreev contact, which provides better performance. This concept allows for better matching with the junction gate field-effect transistor (JFET) readout, suppresses charging noise related to the Coulomb blockade due to the small area of tunnel junctions and decreases the volume of a normal absorber for further improvement of the noise performance. The reliability of a 2D array is considerably increased due to the parallel and series connections of many CEBs.
Estimations of the CEB noise with JFET readout give an opportunity to realise a noise equivalent power (NEP) that is less than photon noise, specifically, NEP = 4 10−19 W/Hz1/2 at 7 THz for an optical power load of 0.02 fW.
- Cold-electron bolometer
- 2D array
- Focal plane antenna
- SIN tunnel junction
- Andreev contact
- JFET readout
- 74.50. + r
Cosmic microwave background (CMB) measurements were ranked second by the journal Science among the top 10 Achievements of the Decade. In 2000 and 2003, an experiment known as Balloon Observations of Millimetric Extragalactic Radiation and Geophysics (BOOMERanG) measured the CMB in detail in patches of the sky. Then in 2003, NASA's space-based Wilkinson Microwave Anisotropy Probe mapped the CMB across the sky, producing an exquisite “baby” picture of the cosmos. These and the measurements that followed have started to transform cosmology from a largely qualitative endeavour to a precision science with a standard theory, known as Precision Cosmology[1, 2]. Recent cosmology experiments have discovered that the universe mainly consists of dark energy and dark matter. Several cosmology instruments (e.g. BOOMERanG-3, SPICA (Space Infra-Red Telescope for Cosmology and Astrophysics telescope), MILLIMETRON, and B-POL) are currently being designed to resolve the nature of these dark components. The science of SPICA and MILLIMETRON will include the cool universe, encompassing galaxy formation in the early distant universe, star formation in the local universe and planet formation in our own galaxy. The observing capabilities on the aforementioned instruments will extend to wavelengths shorter than those on Herschel.
The experiments will require a new generation of ultra-sensitive detectors with exceptionally low noise equivalent power (NEP) in the range of 10−19–10−20 W Hz−1/2. Superconducting detectors with nanoscale absorbers are the most promising candidates for the next generation of supersensitive bolometers. At present, the most widespread superconducting bolometer is a transition-edge sensor (TES)[5, 6]. However, due to artificial DC bias heating, the TES has increased temperature of operation, excess noise and strictly limited saturation power.
The novel concept of the cold-electron bolometers (CEB)[7, 8] has been invented to overcome these problems. The CEB concept is based on a unique combination of the RF (radio frequency) capacitive coupling of the absorber to the antenna through the capacitance of the superconductor-insulator-normal (SIN) tunnel junctions and the direct electron cooling of the absorber by the same SIN tunnel junction. The noise properties of this device are improved considerably by decreasing the electron temperature. Direct electron cooling leads to a considerable increase in the saturation power due to the removal of the incoming power from the sensitive nanoabsorber. The direct electron cooling also provides strong negative electrothermal feedback for the signal, analogous to the TES; but here, the artificial DC heating is replaced by direct electron cooling from the SIN tunnel junctions[10–12] to a minimum temperature (possibly less than phonon temperature). This concept can lead to a major breakthrough in the realisation of supersensitive detectors.
The CEB can be used with both SQUID readout[7–9, 13–15] and JFET/CMOS (Complementary metal–oxide–semiconductor)[9, 14–16]. The JFET and CMOS readouts (with multiplexing) have been used in Herschel and BOOMERanG with high-ohmic Ge boometers. The SQUID readout (with multiplexing) is in a stage of development with low-ohmic TES. The overall goal is to achieve, with a CEB readout from JFET or SQUID (Superconducting quantum interference device), an NEP that is less than the photon noise with a low optical power load for SAFARI (SpicA FAR-infrared instrument) and MILLIMETRON.
2D array of the CEB with a double polarisation cross-dipole antennae
Detection using CEB is obtained by allowing the incoming signal to pass from the antenna to the absorber through the capacitance of an SIN tunnel junction and through an SN Andreev contact. Using this optimal concept of CEB, we achieve several advantages: resistance is decreased for better matching with JFET, the Coulomb blockade is suppressed due to absence of the second junction creating SET transistor and effective volume of the absorber is decreased due to proximity effect of the Andreev contact.
The voltage response is measured by a JFET amplifier in a current-biased mode. The main readout purpose of the 2D array is to match the total dynamic resistance of the array to the noise impedance of a JFET (approximately 0.6 MΩ). Probability of fail is considerably decreased due to parallel combination of several CEBs. For analysed array, we use four CEBs in parallel (W = 4) and eight rows of CEBs in series (N = 8).
Further performance improvements are achieved by placing four bias resistors on chips near the array of bolometers in order to decrease the noise from current-biased resistors and protect the scheme from interferences. RF matching is realised by the resistance of a normal absorber, which is independent of the tunnel junction parameters.
is the cooling thermal conductance of the SIN junction, which provides some electron cooling and helps to avoid the overheating of the absorber.
Due to this correlation, the shot noise is increased by 30–50% in contrast to a CEB in voltage-biased mode, where strong anti-correlation decreases the shot noise.
Estimations were made for the 7 THz channel of SPICA.
The main progress in matching with a JFET readout is achieved by the proper selection of a parallel and series combination of CEBs and a proper resistance of SIN junctions (Figure4). Some progress in NEP is realised due to the replacement of two SIN junctions with one SIN and one SN Andreev contact (Figure3). Better performance is also achieved due to the proper delta suppression of the top superconducting electrode to the level of 400 mK (instead of 1.2 K for clean Al) (Figure4). Overall, the internal overheating of the CEBs by the applied voltage is decreased with arrays of any size (even with N > 100). It is possible that the available CMOS readout system with multiplexing from PACS or JFET multiplexer can be used for matching with the CEB array.
A novel concept of a 2D array of CEBs with double polarised cross-dipole antennas is proposed for spectrometer applications demanding wideband RF matching. A distributed double-polarised dipole antenna with a 2D array of CEBs inserted in the nodes of the antennas is sensitive to both components of the RF signal. The optimal concept of the CEB including a SIN tunnel junction and an SN Andreev contact is used for better performance. This concept has several advantages: it helps to realise better matching with the JFET readout, it suppresses noise related to the Coulomb blockade due to the small area of the tunnel junctions and decreases the volume of a normal absorber for further improvement in noise performance. Besides this, the reliability of a 2D array is considerably increased due to parallel and series connections of CEBs. Estimations of the CEB noise with a JFET readout have shown an opportunity to realise an NEP that is less than photon noise at 7 THz for an optical power load of 0.02 fW.
The author did not provide this information.
The author thanks Phillip Mauskopf, Peter Day, Dmitry Golubev and Mikhail Tarasov for their useful discussions. This work was supported by SNSB, STINT and SI Swedish grants.
- Cho A: Insights of the decade: precision cosmology. Science 2010, 330: 1615. 10.1126/science.330.6011.1615View ArticleGoogle Scholar
- Masi S, et al.: Instrument, method, brightness and polarisation maps from the 2003 flight of BOOMERanG. Astron Astrophys 2006, 458: 687–716. astro-ph/0507509 astro-ph/0507509 10.1051/0004-6361:20053891View ArticleGoogle Scholar
- SPICA: FIR imaging spectrometer-SAFARI (SpicA FAR-infrared Instrument), JAXA-ESA. http://sci.esa.int/science-e/www/area/index.cfm?fareaid=105
- The Space Observatory Millimetron http://www.asc.rssi.ru/millimetron/eng/millim_eng.htm
- Irwin K: An application of electrothermal feedback for high-resolution cryogenic particle-detection. Appl Phys Lett 1998, 1995: 66.Google Scholar
- Lee A, Richards P, Nam S, Cabrera B, Irwin K: A superconducting bolometer with strong electrothermal feedback. Appl Phys Lett 1801, 1996: 69.Google Scholar
- Kuzmin L: Capacitively coupled hot-electron microbolometer as perspective IR and sub-mm wave sensor. In Proc. of the 9th International Symposium on Space Terahertz Technology. Pasadena, 1998: 99–103; Physica B: Condensed Matter; 2000:284–2129.Google Scholar
- Kuzmin L: Optimization of the hot-electron bolometer for space astronomy.SNED Proc., May 2001; Naples. 2001: 145–154. Kuzmin L, Golubev D, Physica C 2002, 378: 372–376.Google Scholar
- Kuzmin L: Ultimate cold-electron bolometer with strong electrothermal feedback. SPIE Proc 2004, 5498: 349–361. 10.1117/12.554317View ArticleGoogle Scholar
- Nahum M, Eiles T, Martinis J: Electronic microrefrigerator based on a NIS tunnel junction. Appl Phys Lett 1994, 65: 3123–3125. 10.1063/1.112456View ArticleGoogle Scholar
- Kuzmin L, Devyatov I, Golubev D: “Cold-electron bolometer” with electronic microrefrigeration and the general noise analysis. Proc SPIE 1998, 3465: 193–199. 10.1117/12.331165View ArticleGoogle Scholar
- Golubev D, Kuzmin L: Nonequilibrium theory of the hot-electron bolometer with NIS tunnel junction. J Appl Phys 2001, 89: 6464–6472. 10.1063/1.1351002View ArticleGoogle Scholar
- Kuzmin L: Ultimate cold-electron bolometer with SIN tunnel junction and Andreev contact. Proc of the 17th Int. Symp. On Space Terahertz Technol., ISSTT 2006, 183–186.Google Scholar
- Kuzmin L: A superconducting cold-electron bolometer with SIS' and Josephson tunnel junctions. J Low Temp Phys 2008, 151: 292–297. 10.1007/s10909-007-9654-9View ArticleGoogle Scholar
- Kuzmin L: Distributed Antenna-Coupled Cold-Electron Bolometers for Focal Plane Antenna. Proc. ISSTT 2008, 154–158.Google Scholar
- Kuzmin L, Mauskopf P, Golubev D: Superconducting cold-electron bolometer with JFET readout for OLIMPO balloon telescope. J Phys Conf Ser (JPCS) 2006, 43: 1298–1302.View ArticleGoogle Scholar
- Day P, LeDuc H, Goldin A, Dowell C, Zmuidzinas J: Far-infrared/summillimeter imager-polarimeter using distributed antenna-coupled transition edge sensors. SPIE Proc 2004, 5498: 857–865. 10.1117/12.552440View ArticleGoogle Scholar
- Kuzmin L, Likharev K: Direct observation of the correlated discrete single-electron tunneling. Jpn J Appl Phys 1987, 26(suppl. 3):1387.View ArticleGoogle Scholar
- Kuzmin L: Array of cold-electron bolometers with SIN tunnel junctions for cosmology experiments. J Phys Conf Ser (JPCS) 2008, 97: 012310.View ArticleGoogle Scholar
- Benoit A, Yates SJC, Grémion E, Camus P, Durand T, Hoffmann C, Leclercq S, Monfardini A, Cavanna A, Etienne B, et al.: A time domain multiplexer for large arrays of high impedance low temperature bolometers. J Low Temp Phys 2008, 151: 940–945. 10.1007/s10909-008-9770-1View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.