- Nano Express
- Open Access
A vacuum-sealed miniature X-ray tube based on carbon nanotube field emitters
© Heo et al.; licensee Springer. 2012
- Received: 13 November 2011
- Accepted: 21 February 2012
- Published: 17 May 2012
A vacuum-sealed miniature X-ray tube based on a carbon nanotube field-emission electron source has been demonstrated. The diameter of the X-ray tube is 10 mm; the total length of the tube is 50 mm, and no external vacuum pump is required for the operation. The maximum tube voltage reaches up to 70 kV, and the X-ray tube generates intense X-rays with the air kerma strength of 108 Gy·cm2 min−1. In addition, X-rays produced from the miniature X-ray tube have a comparatively uniform spatial dose distribution.
- Insertion Position
- Braze Process
- Electron Beam Size
- Spatial Dose Distribution
- Alumina Ceramic Tube
A miniature X-ray tube is a small X-ray generation device generally with a diameter of less than 10 mm [1–5]. Because of the feasible installation in a spatially constrained area and the possibility of electrical on/off control, miniature X-ray tubes can be widely used for nondestructive X-ray radiography, handheld X-ray spectrometers [1, 2], electric brachytherapy, and interstitial or intracavitary radiation therapy or imaging with the substitution of radioactive isotopes [3–5]. Miniature X-ray tubes have been developed mostly using thermionic electron sources [3, 4] or secondary X-ray emission .
Meanwhile, X-ray tubes based on carbon nanotube (CNT) field-emission electron sources have been extensively developed because CNT emitters have several advantages compared with thermionic electron sources. The advantages of CNT emitters include (1) cold electron sources, and hence, little heat is generated inside the tube  which is important for the minimization of an X-ray tube; (2) simplicity and easy controllability in a pulse operation [7, 8]; (3) high current density for electron and X-ray microscopy devices [9, 10]. Several types of X-ray tubes have also been developed using CNT field emitters [11–15]. However, the miniature X-ray tubes are mostly not vacuum-sealed and thus should be operated in a vacuum chamber or with a vacuum pump. In addition, the maximum operating voltages of the miniature X-ray tubes were less than 30 kV. As a consequence, the X-ray tubes have limited practical applications.
In this paper, we report that we have developed a vacuum-sealed miniature X-ray tube using a CNT field emitter. The miniature X-ray tube can be operated up to 70 kV and produces X-rays with very high intensities and a comparatively uniform spatial distribution.
Fabrication of the miniature X-ray tube
Figure 1c,d show a photograph and an X-ray radiograph (XR) of the fabricated miniature X-ray tube, which show the exterior and the interior of the tube, respectively. The diameter of the X-ray tube is 10 mm, and the total length is 50 mm. The weight of the tube is only 14.5 g. All of the connection parts of the X-ray tube are tightly vacuum-sealed. The both ends of the alumina ceramic tube were vacuum-brazed with a focusing electrode assembly and a connecting anode, respectively. Both electrodes were made of Kovar (Carpenter Technology Corporation, Reading, PA, USA) that has a similar thermal expansion coefficient to alumina. The connecting anode was used to interconnect a ceramic tube and a Be X-ray window that have different thermal expansion coefficients. The connecting anode and the Be window were also vacuum-brazed. All the components of the X-ray tube were baked at 550°C for 10 h, and subsequently, these were brazed through a single-step brazing process at 680°C for 30 min in a vacuum furnace. Before the brazing process, electron emission and transport tests of the X-ray tube have been carried out inside a vacuum chamber. The position of the cathode inside the focusing electrode could be finely controlled through this process. A non-evaporable getter film was installed around the focusing electrode to evacuate the X-ray tube. The getter was activated during the brazing process. The outer part of the sealed X-ray tube, except the target, was covered with a layer of silicone resin to improve high-voltage insulation between the cathode and the X-ray target. We observed that the fabricated X-ray tube was stably operated up to 70 kV without any high-voltage breakdown or discharge at both the inner vacuum side and the outer air side.
Performance and characterization of the X-ray tube
Figure 2b shows the dose rate of X-ray that is produced from the miniature X-ray tube and the stability of the dose rate with time. The dose rate was measured with an ionization chamber and an electrometer (PTW 34013 and Unidos-E both from PTW, Freiburg, Germany) at 1 cm apart from the X-ray tube in air. The air kerma strength of the X-ray tube operating at 50 kV with the tube current of 252 μA was as high as 108.1 Gy·cm2 min−1, which is approximately 15 times higher than that of a 10-Ci HDR 192Ir radioisotope source  that is widely used for brachytherapy. The fluctuation of the X-ray dose rate was as low as ±2.7%. In addition, the X-ray tube has worked for over 2 months with no significant change in the electron beam current (0.25 mA at 50 kV) and the X-ray dose rate. The X-ray tube has been continuously operated for approximately 1 h/day. Consequently, the developed miniature X-ray tube produces high enough X-ray output and exhibits very good short-term and long-term stabilities.
To further evaluate the performance of the miniature X-ray tube, the X-ray focal spot size was measured following the European standard EN 12543–5. An XR image of a copperplate (thickness 0.1 mm) was taken with a magnification factor of 1.25 (Figure 4b). The image profiles in both the horizontal and vertical directions were analyzed, and from the analysis, the X-ray focal spot size was calculated to be 3.72 mm in the horizontal direction and 3.64 mm in the vertical direction. The focal spot size of the X-ray corresponds to the electron beam size at the X-ray target. Therefore, the measurement results for the X-ray focal spot size suggest that the inner diameter (7 mm) of the present miniature X-ray tube can be reduced, and accordingly, the X-ray tube can be further minimized.
In summary, we have demonstrated a vacuum-sealed miniature X-ray tube based on CNT field-emission electron source. The X-ray tube can be operated up to 70 kV, and high-dose X-rays are generated with a comparatively good spatial dose distribution. Due to the small electron beam size in the X-ray tube, the prototype X-ray tube can be further miniaturized. We believe that such a vacuum-sealed miniature X-ray tube can be used for various industrial and medical diagnostic/therapy purposes.
SHH has a Ph.D. in Nuclear Engineering and is a president of a ventured company. HJK and JMH are MS students. SOC is a professor of Nuclear Engineering.
This study was supported by the National Research Foundation of Korea (NRF) grant funded by the Korean government (MEST) (no. 2011–0018418) and the R&D Program of MKE/KEIT (10035553).
- Lovoi P, Asmus J: An X-ray microprobe for in-situ stone and wood characterization. In Lasers in the Conservation of Artworks. LACONA V Proceedings: September 15–18, 2003; Osnabrück. Edited by: Dickmann K, Fotakis C, Asmus J. Heidelberg: Springer; 2005:353–356.View ArticleGoogle Scholar
- Koppel L, Marshall J: A miniature metal-ceramic x-ray source for spacecraft instrumentation. Rev Sci Instrum 1998, 69: 1893–1897. 10.1063/1.1148860View ArticleGoogle Scholar
- Dickler A: Xoft Axxent (R) electronic brachytherapy-a new device for delivering brachytherapy to the breast. Nat Clin Pract Oncol 2009, 6: 138–142.View ArticleGoogle Scholar
- Schneider F, Fuchs H, Lorenz F, Steil V, Ziglio F, Kraus-Tiefenbacher U, Lohr F, Wenz F: A Novel device for intravaginal electronic brachytherapy. Int J Radiat Oncol Biol Phys 2009, 74: 1298–1305. 10.1016/j.ijrobp.2009.01.082View ArticleGoogle Scholar
- Gutman G, Strumban E, Sozontov E, Jenrow K: X-ray scalpel - a new device for targeted x-ray brachytherapy and stereotactic radiosurgery. Phys Med Biol 2007, 52: 1757–1770. 10.1088/0031-9155/52/6/015View ArticleGoogle Scholar
- Mahapatra D, Sinha N, Yeow J, Melnik R: Field emission from strained carbon nanotubes on cathode substrate. Appl Surf Sci 2008, 255: 1959–1966. 10.1016/j.apsusc.2008.06.128View ArticleGoogle Scholar
- Liu Z, Yang G, Lee Y, Bordelon D, Lu J, Zhou O: Carbon nanotube based microfocus field emission x-ray source for microcomputed tomography. Appl Phys Lett 2006, 89: 103111. 10.1063/1.2345829View ArticleGoogle Scholar
- Zhang J, Yang G, Cheng Y, Gao B, Qiu Q, Lee Y, Lu J, Zhou O: Stationary scanning x-ray source based on carbon nanotube field emitters. Appl Phys Lett 2005, 86: 184104. 10.1063/1.1923750View ArticleGoogle Scholar
- de Jonge N, Lamy Y, Schoots K, Oosterkamp T: High brightness electron beam from a multi-walled carbon nanotube. Nature 2002, 420: 393–395. 10.1038/nature01233View ArticleGoogle Scholar
- Heo S, Ihsan A, Cho S: Transmission-type microfocus x-ray tube using carbon nanotube field emitters. Appl Phys Lett 2007, 90: 183109. 10.1063/1.2735549View ArticleGoogle Scholar
- Haga A, Senda S, Sakai Y, Mizuta Y, Kita S, Okuyama F: A miniature x-ray tube. Appl Phys Lett 2004, 84: 2208–2210. 10.1063/1.1689757View ArticleGoogle Scholar
- Senda S, Sakai Y, Mizuta Y, Kita S, Okuyama F: Super-miniature x-ray tube. Appl Phys Lett 2004, 85: 5679–5681. 10.1063/1.1832733View ArticleGoogle Scholar
- Tan T, Sim H, Lau S, Yang H, Tanemura M, Tanaka J: X-ray generation using carbon-nanofiber-based flexible field emitters. Appl Phys Lett 2006, 88: 103105. 10.1063/1.2182022View ArticleGoogle Scholar
- Cao G, Burk L, Lee Y, Calderon-Colon X, Sultana S, Lu J, Zhou O: Prospective-gated cardiac micro-CT imaging of free-breathing mice using carbon nanotube field emission x-ray. Medical Physics 2010, 37: 5306–5312. 10.1118/1.3491806View ArticleGoogle Scholar
- Sprenger F, Calderon-Colon X, Gidcumb E, Lu J, Qian X, Spronk D, Tucker A, Yang G, Zhou O: Stationary digital breast tomosynthesis with distributed field emission x-ray tube. Proc SPIE 2011, 7961: 79615I.View ArticleGoogle Scholar
- Heo S, Ihsan A, Yoo S, Ali G, Cho S: Stable field emitters for a miniature x-ray tube using carbon nanotube drop drying on a flat metal tip. Nanoscale Res Lett 2010, 5: 720–724. 10.1007/s11671-010-9537-xView ArticleGoogle Scholar
- Herrmannsfeldt WB, Herrmannsfeldt GA: EGN Electron Optics Program. California: SLAC, Stanford; 1993.Google Scholar
- Ihsan A, Heo S, Kim H, Kang C, Cho S: An optimal design of X-ray target for uniform X-ray emission from an electronic brachytherapy system. Nucl Instrum Methods Phys Res, Sect B 2011, 269: 1053–1057. 10.1016/j.nimb.2011.03.001View ArticleGoogle Scholar
- Ihsan A, Heo S, Cho S: Optimization of X-ray target parameters for a high-brightness microfocus X-ray tube. Nucl Instrum Methods Phys Res, Sect B 2007, 264: 371–377. 10.1016/j.nimb.2007.09.023View ArticleGoogle Scholar
- Nath R, Anderson L, Luxton G, Weaver K, Williamson J, Meigooni A: Dosimetry of interstitial brachytherapy sources: recommendations of the AAPM Radiation-Therapy Committee Task Group no. 43. Medical Physics 1995, 22: 209–234. 10.1118/1.597458View ArticleGoogle Scholar
- Bearden J: X-ray wavelengths. Reviews of Modern Physics 1967, 39: 78–124. 10.1103/RevModPhys.39.78View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.