Towards on-chip time-resolved thermal mapping with micro-/nanosensor arrays
© Liu et al.; licensee Springer. 2012
Received: 16 July 2012
Accepted: 25 July 2012
Published: 29 August 2012
In recent years, thin-film thermocouple (TFTC) array emerged as a versatile candidate in micro-/nanoscale local temperature sensing for its high resolution, passive working mode, and easy fabrication. However, some key issues need to be taken into consideration before real instrumentation and industrial applications of TFTC array. In this work, we will demonstrate that TFTC array can be highly scalable from micrometers to nanometers and that there are potential applications of TFTC array in integrated circuits, including time-resolvable two-dimensional thermal mapping and tracing the heat source of a device. Some potential problems and relevant solutions from a view of industrial applications will be discussed in terms of material selection, multiplexer reading, pattern designing, and cold-junction compensation. We show that the TFTC array is a powerful tool for research fields such as chip thermal management, lab-on-a-chip, and other novel electrical, optical, or thermal devices.
One of the major bottlenecks of fast development in the modern electronics industry is heat. Managing the high heat flux generated by billions of transistors in an integrated circuit (IC) is the main challenge. A scalable, built-in, and agile temperature sensor which can provide real-time thermal history of an IC chip is essential to thermal failure analysis . However, thermal measurement at microscale and nanoscale is a technical challenge. When both high spatial resolution and high temperature sensitivity are required, only a few approaches, such as scanning thermal microscopy , luminescent nanoparticles, and infrared thermography [3–5], remain in the list of suitable candidates. If built-in thermal sensors are required for time-resolvable solutions in a solid device, such as in IC chips, micro-electro-mechanical system, a lab-on-a-chip, or a flexible printed circuits, then it seems that only stripe-shaped resistive sensor  and thin-film thermocouple (TFTC) are left in the candidate list. The passive nature of a thermocouple makes it superior to a resistive thermal sensor (usually made of metal or semiconductors) because a thermocouple provides a far more rapid response, does not require external excitation voltage or current, and generates no additional heat which may cause unexpected electrical noise and error in temperature measurement.
where S α and S β represent the Seebeck coefficients of α and β, and Th and Tc are the temperatures at the hot end and cold end, respectively. TP means the thermopower of thermocouple, which is the sensitivity of the sensor. Given the measured ΔV Tc, and TP, one can easily calculate the Th. By embedding thermocouples in an electrical device, e.g., an IC chip, one can monitor the temperature at the area of thermocouple junction even after packaging. In addition, the measured voltage signal is just relative to the temperature of the junction area; thus, by shrinking the junction area of the thermocouple, one can obtain a highly localized temperature measurement. Pioneering work on scaling down the size of a TFTC has pushed its junction area to 100 × 100 nm , suggesting the potential for local temperature sensing at the nanoscale. Finally, the fabrication process for a TFTC is compatible with the well-developed microelectronic techniques. As a result, TFTCs have attracted much attention in recent years [9–15].
In 2005, Park et al. firstly constructed 100 TFTCs into an array providing two-dimensional (2D) temperature mapping in an area of around a square centimeter . A few years later, Grayson et al. discussed some considerations in the progress of instrument TFTC array into real IC chips [13, 14]. Recently, we presented a novel method for time-resolved 2D mapping by TFTC array  and firstly constructed a functional TFTC array made of a single material . As a built-in solution, TFTC array is gaining more and more attention. However, to further push forward the development of TFTC array and its instrumentation in electronics industry, more potential advantages have to be explored, and some related issues have to be addressed.
In this work, we report on our recent work on TFTC array, including construction of the first nanoscale TFTC array in an area of 5 × 5 μm. We demonstrate that besides real-time temperature mapping, the TFTC array can also be utilized in mapping time-resolved heat flux and tracing a possible hot spot on a surface. We also discuss some key issues of instrumentation in IC chips in the future, including choice of material, multiplexing, pattern designing, and cold end problems.
Typical physical properties and figure of merits of candidate materials (300 K)
1.27E − 7
8.9E − 3
7E − 8
1.5E − 2
2.65E − 8
1.4E − 4
3.3E − 7
8.9E − 6
8.1E − 6
3.8E − 3
2.2E − 5
4.6E − 3
2.4E − 5
7.9E − 3
1.9E − 5
5.0E − 3
Moreover, unlike the intrinsic Seebeck coefficient of metals, the Seebeck coefficient of semiconductor is highly extrinsic in relevance with the doping density, crystal structure, and resistivity. For example, the Seebeck coefficient of p-type silicon varied from 300 to 1,600 μV/K at room temperature . Because of its relevance to fabrication process, the thermopower of semiconductor-based thermocouple may vary from batch to batch or even from one to another, which was hard to put into industrial batch productions. As for the metal-based thermocouples, they were put into wide applications for many years, and industrial standard thermocouples were all made of metals or alloys. Thin-film metal thermocouple can generate a similar thermopower given similar film thicknesses. Varrenti et al. first reported the reproducibility of Cr-Ni TFTCs, both inter- and intra-batch . Our work also contributed to the high stability of Cr-Ni TFTCs by massive calibrations .
As to the fabrication, Cr-Ni TFTCs can also be produced in CMOS-compatible processes just like the silicon-based temperature sensors. The differences were that sputtering was used in metal fabrications, and diffusion or ion implantation was adopted in doped silicon stripe fabrication. Si-based thermocouple fabrication may require additional lithography and etching to remove the superfluous undoped silicon to avoid short circuit. Besides, metals can be incorporated in different layers of ICs. The silicon-based device always synthesized in the same layer of the ICs, consuming precious spaces.
Array designing and multiplexing
As to on-chip applications, a high surface density of TFTCs certainly increases the spatial resolution and accuracy in temperature measurement, but it also remarkably increases the number of pads and connections for the TFTC array, leading to complex interconnection. Therefore, the density of sensors in the array should be balanced between being enough to give required sensing resolution and being minimum to reduce the complexity of interconnection and space of pads.
We have found that matrices with relatively high densities were not necessary in integrated circuit temperature measurement since the ‘hot spot’ in a chip will provide temperature at least 20 times higher than the ambient temperature. Matrices of 3 × 3 or 3 × 6 TFTC array provided a temperature mapping good enough to help in judging where the hot spot was and how the thermal field time-dependently changed.
In addition, the density of TFTC array may have an influence on the real-time monitoring of local temperatures. Usually, one uses only one voltmeter to subsequently measure all the thermocouples in one array, with the help of a multiplexer. However, since a reliable reading process by a voltmeter needs a minimum time of 1 to 10 ms, if the total number of sensors in a TFTC array is too large, then one cycle of the measurements which ensures each individual sensor is measured once may take too long; thus, a real-time monitoring of the 2D temperature map may not be practical.
Results and discussion
Thermal 2D mapping
Tracing the position of heat source
In on-chip applications, one cannot expect the hot spot to be always right on top of a TFTC. Unexpected failure in a working chip is always accompanied with unexpected hot spot induced. Tracing the possible locations of these unexpected hot spots is tricky for technicians. Inspired by a similar subject in signal processing, we propose to use a regular TFTC array to trace the heat.
Industrial on-chip applications
Another potential problem is that the temperature rises in the cold ends of TFTCs. In a TFTC built-in IC chip, because of the excellent thermal conductivity of silicon substrate when the IC circuit has been run for a long time, joule heat generated from the running current may heat up the entire chip, causing a remarkable rise of temperature at the cold ends of the TFTCs, thus causing a huge error in temperature readings. It seems that in this case, on-chip sensors of diodes or transistors can provide better temperature reading . However, there are at least four defects which limit the development of these methods and why many endeavors were made in developing new ways of temperature sensing in IC chips.
Firstly, such methods all require active parts like electrical sources, which bring additional consumption and self-induced heat. Secondly, diode, transistors, and thermistor sensors often exhibit nonlinear temperature characteristics in operating temperature range, and their property may vary sensitively to the fabrication process, so calibrations will be often adopted to test the temperature characteristic. Offset was also usually programmed into the sensors to give additional compensation. Thirdly, the diode sensors cannot be placed just in the hottest area of the chip because of routing and I/O limitations . In chips with elements of high density like the Intel® CoreTM Duo processor, the hot spots tend to vary from place to place according to different loaded watt power. Imprecision from 2°C to 10°C caused by non-linearity and hot spot drifting was not strange by adopting diode temperature sensors .
In contrast, although the inherent defects of thermocouple-based sensor can just measure the temperature differences spanned on the device, they have all the merits of having no active parts, no electrical sources, being simple and cheap, having linear responses, and being repeatable and quick enough for time-resolved profile. Possible solutions to minimize the impact of temperature of cold end were addressed as follows:
It is true that the difficulty of local temperature mapping by TFTC array increases as the chip area is scaling down, especially for very small silicon-based chips, e.g., 5 × 5 mm. It will not be easy to carry out on-chip thermal management based on the TFTC array. However, one of the properties of TFTCs is that their property will not be affected by the device routing, and the output signal of TFTCs is only contingent with temperature differences between the hot ends and the cold ends. Thus, we can arrange a common cold end for all the TFTCs. As they are sharing one cold end, we can still map the relative temperature differences and locate the hot spot. In most situations, IC chips which need thermal management will not be as small as 5 × 5 mm, and the hot spots are usually distributed locally in only parts of the working components of high density , so the distance between the junction area and the cold end can easily exceed 1 cm, and arranging the cold end on the fringe of the silicon die will not provide significant measurement error. A cooling base or cooling fan will soon dissipate the heat at the marginal area of the chip. We have produced micro- or nano-TFTC arrays on silicon substrates from 1 × 1 cm to 4 inch in diameter, with pads on the wafer edges. The TFTC array still functioned well by simply referring the temperature of the pads as room temperature.
We have demonstrated micro- and nanoscale TFTC arrays as a powerful built-in thermal mapping tool. Using multiplexing technique, time-resolved temperature mapping can be easily drawn from TFTC array readings. Moreover, we have shown that TFTC array can also map real-time heat flux in plane, and the 2D mapping results can be used to precisely trace the heating source located in the surface.
Bearing in mind that the TFTC array can be used for on-chip solutions in the future, we discussed several remaining issues for its industrialization, i.e., choices of materials, density of sensor, sampling rate, common-lead strategy, and cold end temperature shift and solutions. We have demonstrated that a configuration of common lead can substantially reduce the number of pads by 50%.
We conclude that a variety of configuration of TFTC arrays is the most promising candidate as built-in sensors in 2D mapping of local temperature for solid devices such as IC chips because they are highly scalable from macro- to nanoscale, providing the potential of extremely high resolution and because they have a simple configuration, passive working mode, time-resolvable capability for real-time monitoring, and the capability for versatile applications.
This work was financially supported by NSFC (grant no. 11074010) and MOST of China (grant 2011CB933002).
- Altet J, Claeys W, Dilhaire S, Rubio A: Dynamic surface temperature measurements in ICs. Proc IEEE 2006, 94: 1519.View ArticleGoogle Scholar
- Williams C, Wickramasinghe H: Scanning thermal profiler. Appl Phys Lett 1986, 49: 1587. 10.1063/1.97288View ArticleGoogle Scholar
- Brites C, Lima P, Silva N, Millán A, Amaral V, Palacio F, Carlos L, Brites C: A luminescent molecular thermometer for long-term absolute temperature measurements at the nanoscale. Adv Mater 2010, 22: 4499. 10.1002/adma.201001780View ArticleGoogle Scholar
- Gota C, Okabe K, Funatsu T, Harada Y, Uchiyama S: Hydrophilic fluorescent nanogel thermometer for intracellular thermometry. J Am Chem Soc 2009, 131: 2766. 10.1021/ja807714jView ArticleGoogle Scholar
- Lee DH: Thermal analysis of integrated-circuit chips using thermographic imaging techniques. IEEE Trans Instrum Meas 1994, 43: 824. 10.1109/19.368060View ArticleGoogle Scholar
- Li DY, Wu YY, Kim P, Shi L, Yang PD, Majumdar A: Thermal conductivity of individual silicon nanowires. Appl Phys Lett 2003, 83: 2934. 10.1063/1.1616981View ArticleGoogle Scholar
- Jang M, Park Y, Jun M, Hyun Y, Choi S, Zyung T: The characteristics of Seebeck coefficient in silicon nanowires manufactured by CMOS compatible process. Nanoscale Res Lett 2010, 5: 1654. 10.1007/s11671-010-9690-2View ArticleGoogle Scholar
- Chu DC, Wong W, Goodson K, Pease R: Transient temperature measurements of resist heating using nanothermocouples. J Vac Sci Technol B 2003, 21: 2985. 10.1116/1.1624255View ArticleGoogle Scholar
- Szentpáli B, Basa P, Fürjes P, Battistig G, Bársony I, Károlyi G, Berceli T, Rymanov V, Stöhr A: Thermopile antennas for detection of millimeter waves. Appl Phys Lett 2010, 96: 133507. 10.1063/1.3374445View ArticleGoogle Scholar
- Salvadori M, Vaz A, Teixeira F, Cattani M, Brown I: Thermoelectric effect in very thin film Pt/Au thermocouples. Appl Phys Lett 2006, 88: 133106. 10.1063/1.2189192View ArticleGoogle Scholar
- Duby S, Ramsey B, Harrison D: Printed thick-film thermocouple sensors. Electron Lett 2005, 41: 312. 10.1049/el:20057988View ArticleGoogle Scholar
- Park J, Taya M: Micro-temperature sensor array with thin-film thermocouples. Electron Lett 2004, 40: 599. 10.1049/el:20040387View ArticleGoogle Scholar
- Long JY, Klock A, Zhou CL, Memik S, Grayson M: Proc 16th THERMINIC. 16th International Workshop on Thermal Investigations of ICs and Systems, September 2010. In IOTA: towards an integrated on-chip thermocouple array. Edited by: Altet J, Courtois B, Rencz M, Rubio A. EDA Publishing, Barcelona; 2010.Google Scholar
- Varrenti A, Zhou CL, Klock A, Chyung S, Long JY, Memik S, Grayson M: Thermal sensing with lithographically patterned bimetallic thin-film thermocouples. IEEE Electron Dev Lett 2011, 32: 818.View ArticleGoogle Scholar
- Liu HX, Sun WQ, Chen Q, Xu SY: Thin-film thermocouple array for time-resolved local temperature mapping. IEEE Electron Dev Lett 2011, 32: 1606.View ArticleGoogle Scholar
- Liu HX, Sun WQ, Xu SY: An extremely simple thermocouple made of a single layer of metal. Adv Mater 2012, 24: 3275. 10.1002/adma.201200644View ArticleGoogle Scholar
- Fulkerson W, Moore J, Williams R, Graves R, McElroy D: Thermal conductivity, electrical resistivity, and Seebeck coefficient of Silicon from 100 to 1300 K. Phys Rev 1968, 167: 765. 10.1103/PhysRev.167.765View ArticleGoogle Scholar
- Strasser M, Aigner R, Franosch M, Wachutka G: Miniaturized thermoelectric generators based on poly-Si and poly-SiGe surface micromachining. Sens Actuator A, Phys 2002, 535: 97–98.Google Scholar
- Lahiji G, Wise K: A batch-fabricated silicon thermopile infrared detector. IEEE Tran Electron Dev 1982, 29: 14.View ArticleGoogle Scholar
- Lenggenhager R, Baltes H, Peer J, Forster M: Thermoelectric infrared sensors by CMOS technology. IEEE Electron Dev Lett 1992, 13: 454.View ArticleGoogle Scholar
- Baltes H, Paul O, Brand O: Micromachined thermally based CMOS microsensors. Proc IEEE 1998, 86: 1660. 10.1109/5.704271View ArticleGoogle Scholar
- Herwaarden A, Sarro P: Thermal sensors based on the Seebeck effect. Sens Actuator. 1986, 10: 321. 10.1016/0250-6874(86)80053-1View ArticleGoogle Scholar
- Rencz M, Kollár E, Székely V: Heat-flux sensor to support transient thermal characterisation of IC packages. Sens Actuator A, Phys 2004, 116: 284. 10.1016/j.sna.2004.05.002View ArticleGoogle Scholar
- Han I, Kim S: Diode temperature sensor array for measuring micro-scale surface temperatures with high resolution. Sens Actuator A, Phys 2008, 141: 52. 10.1016/j.sna.2007.07.020View ArticleGoogle Scholar
- Rotem E, Hermerding J, Aviad C, Harel C: Proc 12th THERMINIC. 12th International Workshop on Thermal Investigations of ICs and Systems, September 2006. In Temperature measurement in the Intel® CoreTM Duo processor. Edited by: Courtois B, Rencz M, Lasance C, Szekely V. EDA Publishing, Côte d’Azur; 2006.Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.