Fabrication and NO2 gas sensing performance of TeO2-core/CuO-shell heterostructure nanorod sensors
© Park et al.; licensee Springer. 2014
Received: 26 August 2014
Accepted: 14 November 2014
Published: 27 November 2014
TeO2-nanostructured sensors are seldom reported compared to other metal oxide semiconductor materials such as ZnO, In2O3, TiO2, Ga2O3, etc. TeO2/CuO core-shell nanorods were fabricated by thermal evaporation of Te powder followed by sputter deposition of CuO. Scanning electron microscopy and X-ray diffraction showed that each nanorod consisted of a single crystal TeO2 core and a polycrystalline CuO shell with a thickness of approximately 7 nm. The TeO2/CuO core-shell one-dimensional (1D) nanostructures exhibited a bamboo leaf-like morphology. The core-shell nanorods were 100 to 300 nm in diameter and up to 30 μm in length. The multiple networked TeO2/CuO core-shell nanorod sensor showed responses of 142% to 425% to 0.5- to 10-ppm NO2 at 150°C. These responses were stronger than or comparable to those of many other metal oxide nanostructures, suggesting that TeO2 is also a promising sensor material. The responses of the core-shell nanorods were 1.2 to 2.1 times higher than those of pristine TeO2 nanorods over the same NO2 concentration range. The underlying mechanism for the enhanced NO2 sensing properties of the core-shell nanorod sensor can be explained by the potential barrier-controlled carrier transport mechanism.
61.46. + w; 07.07.Df; 73.22.-f
KeywordsTeO2 nanorods CuO shells Gas sensors Response NO2
In recent years, one-dimensional (1D) nanostructure-based sensors attracted considerable attention owing to their high surface-to-volume ratios [1–5]. Considerable effort has been made to develop 1D nanostructured gas sensors with good sensing performances, but further improvements in the sensitivity of 1D nanostructured sensors are needed. The fabrication of heterostructures [6–8] is a promising technique to improve the sensitivity of the 1D nanostructured sensors. The improved sensing performance of the heterostructured 1D sensors has been attributed to a range of factors including increased potential barriers at the interface of the heterostructure [9, 10], modulated depletion layer [11, 12], band bending due to equilibration of the Fermi energy levels , synergistic surface reactions , etc.
Paratellurite (α-TeO2) is a metal oxide semiconductor with a distorted rutile structure . TeO2 has applications in optical storage, laser devices and gas sensors, dosimeters, modulators, and deflectors owing to its unique properties such as high refractive index and high optical nonlinearity . TeO2-nanostructured sensors have attracted less attention compared to other metal oxide semiconductor materials such as ZnO, In2O3, TiO2, Ga2O3, etc. In 2007, Liu et al.  synthesized TeO2 nanowires that were sensitive to NO2, NH3, and H2S gases. According to their results, TeO2 1D nanostructures are promising for producing low power consumption gas sensors. The incorporation of a surface decoration or heterostructure formation technique can improve their sensing performance further. In this regard, a recent study reported the sensing properties of Pt-doped TeO2 nanorods . On the other hand, this paper reports the synthesis of TeO2-core/CuO-shell nanorods and the sensing properties of multiple networked TeO2-core/CuO-shell nanorod gas sensors toward NO2 gas. The underlying mechanism for the enhanced sensing performance of the core-shell nanorod sensors is also discussed.
TeO2/CuO core-shell nanorods were synthesized using a two-step process: thermal evaporation of Te powder followed by sputter deposition of CuO. TeO2 nanorods were synthesized on a p-type Si (100) substrate in a quartz tube furnace by thermal evaporation of Te powder at 400°C in air without a metal catalyst or the supply of other gas. The thermal evaporation process was conducted at room temperature for 1 h and the furnace was cooled to room temperature. Subsequently, the TeO2 nanorods were coated with a thin CuO layer by sputtering a CuO target by radio frequency (RF) magnetron sputtering from a CuO target. The base and working pressure was 5.0 × 10-6 Torr and 2.0 × 10-2 Torr, respectively, and the N2 gas flow rate was 20 cm3/min throughout the evaporation process. The RF sputtering power and sputtering time were 100 W and 20 min, respectively.
The structure and morphology of the nanorod samples were characterized by scanning electron microscopy (SEM, Hitachi S-4200, Billerica, MA, USA), transmission electron microscopy (TEM, Philips CM-200, Eindhoven, the Netherlands), and selected area electron diffraction. X-ray diffraction (XRD, Philips X’pert MRD, Eindhoven, the Netherlands) patterns were performed using Cu Kα radiation (0.15406 nm). Energy-dispersive X-ray spectroscopy (EDS) was carried out to examine the elemental composition of the core-shell nanorod samples. The resistance of multiple networked pristine TeO2 nanorod and TeO2/CuO core-shell nanorod sensors were measured using a Keithley source meter-2612 at a source voltage of 10 V at 150°C and 50% RH. The 50% relative humidity might be somewhat high for sensing tests. A flow-through technique was used to test the gas sensing properties. NO2 gas diluted with synthetic air at different ratios was injected into the testing tube at a constant flow rate of 200 cm3/min. The detailed procedures for sensor fabrication and the sensing test are reported elsewhere .
Results and discussion
Responses of the TeO 2 /CuO nanorod sensor to NO 2 gas at different concentrations at 150°C
Response (Ra/Rg, %)
Figure 3c compares the response to NO2 gas between pristine TeO2 nanorods and TeO2/CuO core-shell nanorods in the NO2 concentration range below 10 ppm. The response of an oxide semiconductor sensor can be expressed as R = A [C]n + B, where A and B, n, and [C] are constants, exponent, and target gas concentration, respectively . Data fitting gave R = 7.52 [C] + 132.5 and R = 27.48 [C] + 153.9 for the pristine TeO2 nanorod and TeO2-core/CuO-shell nanorod sensors, respectively. The core-shell nanorod sensor showed stronger response and higher increasing rate in response to NO2 gas at lower concentrations than the pristine nanorod sensor.
Comparison of the responses of the TeO 2 /CuO core-shell nanorod sensor with those of other oxide 1D nanostructure sensors
In-doped SnO2 nanoparticles
SnO2 hollow spheres
Ru-doped SnO2 nanowire
WO3-doped SnO2 thin film
Au-doped WO3 powders
Mesoporous WO3 thin film
180 to 300
CdO nanowire (porous)
70 to 2,000
20 to 320
TeO2/CuO core-shell nanorods were synthesized using a two-step process: the synthesis of TeO2 nanorods by thermal evaporation of Te powder and sputter deposition of CuO. The cores and shells of the nanorods were single crystal TeO2 and polycrystalline CuO, respectively. The responses of the TeO2 nanorods to NO2 were improved approximately 2.1- to 2.1-fold at NO2 concentrations of 0.5 to 10 by coating them with CuO. The responses of the core-shell nanorods to NO2 gas were also comparable or superior to those of the other metal oxide semiconductor nanostructured sensors reported previously. The enhanced response of the TeO2/CuO core-shell nanorods to NO2 gas may be due to modulation of the heights of the potential barriers formed at three different places in the multiple networked 1D nanostructure sensor: the TeO2 core-CuO shell interface, the CuO-CuO homojunction at the contact of two core-shell nanorods, and the grain boundaries in the polycrystalline CuO shell layers.
This study was supported by the 2010 Core Research Program through the National Research Foundation of Korea (NRF) funded by the Ministry of Education, Science and Technology.
- Tippins HH: Optical absorption and photoconductivity in the band edge of β-Ga2O3. Phys Rev A 1965, 140: 316–319. 10.1103/PhysRev.140.A316View ArticleGoogle Scholar
- Yamazoe N: New approaches for improving semiconductor gas sensors. Sens Actuators B 1991, 5: 7–19. 10.1016/0925-4005(91)80213-4View ArticleGoogle Scholar
- Gundiah G, Govindaraj A, Rao CNR: Nanowires, nanobelts and related nanostructures of Ga2O3. Chem Phys Lett 2002, 351: 189–194. 10.1016/S0009-2614(01)01372-0View ArticleGoogle Scholar
- Zhang HZ, Kong YC, Wang YZ, Du X, Bai ZG, Wang JJ, Yu DP, Ding Y, Hang QL, Feng SQ: Ga2O3 nanowires prepared by physical evaporation. Sol State Comm 1999, 9: 677–682.View ArticleGoogle Scholar
- Kim BC, Sun KT, Park KS, Im KJ, Noh T, Sung MY, Kim S, Nahm S, Choi YN, Park SS: β-Ga2O3 nanowires synthesized from milled GaN powders. Appl Phys Lett 2002, 80: 479–481. 10.1063/1.1435073View ArticleGoogle Scholar
- Yan C, Le BH, Kang DJ: Ultrasensitive single crystalline TeO2 nanowire based hydrogen gas sensors. J Mater Chem A 2014, 2: 5394–5398. 10.1039/c4ta00135dView ArticleGoogle Scholar
- Sanchez-Castillo MA, Couto C, Kim WB, Dumesic JA: Gold-nanotube membranes for the oxidation of CO at gas-water interfaces. Angew Chem Int Ed 2004, 43: 1140–1142. 10.1002/anie.200353238View ArticleGoogle Scholar
- Jagerszki G, Giurskisanyi RE, Hoefler L, Pretsch E: Hybridization-modulated ion fluxes through peptide-nucleic-acid-functionalized Gold nanotubes. A new approach to quantitative label-free DNA analysis. Nano Lett 2007, 7(6):1609–1612. 10.1021/nl0705438View ArticleGoogle Scholar
- Wang W, Li Z, Zheng W, Huang H, Wang C, Sun J: Cr2O3-sensitized ZnO electrospun nanofibers based ethanol detectors. Sens Actuators B 2010, 143: 754–758. 10.1016/j.snb.2009.10.016View ArticleGoogle Scholar
- Sun P, Sun Y, Ma J, You L, Lu G, Fu W, Li M, Yang H: Synthesis of novel SnO2/ZnSnO3 core-shell microspheres and their gas sensing properties. Sens Actuators B 2011, 155: 606–611. 10.1016/j.snb.2011.01.017View ArticleGoogle Scholar
- Wang L, Kang Y, Wang Y, Zhu B, Zhang S, Huang W: CuO nanoparticle decorated ZnO nanorod sensor for low-temperature H2S detection. Mater Sci Eng C 2012, 32: 2079–2085. 10.1016/j.msec.2012.05.042View ArticleGoogle Scholar
- Liu Y, Zhu G, Chen J, Xu H, Shen X, Yuan A: Co3O4/ZnO nanocomposites for gas-sensing applications. Appl Surf Sci 2013, 265: 379–384.View ArticleGoogle Scholar
- Kusior A, Radecka M, Rekas M, Lubecka M, Zakrzewska K, Reszka A: Sensitization of gas sensing properties in TiO2/SnO2 nanocomposites. Procedia Eng 2012, 47: 1073–1076.View ArticleGoogle Scholar
- De Lacy Costello BPJ, Ewen RJ, Ratcliffe NM, Sivanand PS: Thick film organic vapour sensors based on binary mixtures of metal oxides. Sens Actuators B 2003, 92: 59–166.View ArticleGoogle Scholar
- Kim ID, Rothschild A, Lee BH, Kim DY, Jo SM, Tuller HL: Ultrasensitive chemiresistors based on electrospun TiO2 nanofibers. Nano Lett 2006, 6: 2009–2013. 10.1021/nl061197hView ArticleGoogle Scholar
- Jin C, Park S, Kim H, Lee C: Enhanced gas sensing properties of Pt-loaded TeO2 nanorods. Bull Kor Chem Soc 2012, 33: 1851–1855. 10.5012/bkcs.2012.33.6.1851View ArticleGoogle Scholar
- Li YJ, Li KM, Wang CY, Kuo CI, Chen LJ: Low-temperature electrodeposited Co-doped ZnO nanorods with enhanced ethanol and CO sensing properties. Sens Actuators B 2012, 161: 734–739. 10.1016/j.snb.2011.11.024View ArticleGoogle Scholar
- Jin C, Park S, Kim H, Lee C: Ultrasensitive multiple networked Ga2O3-core/ZnO-shell nanorods gas sensors. Sens Actuators B 2012, 161: 223–228. 10.1016/j.snb.2011.10.023View ArticleGoogle Scholar
- Williams DE: Solid State Gas Sensors. Bristol: Hilger; 1987.Google Scholar
- Oh E, Choi HY, Jung SH, Cho S, Kim JC, Lee KH, Kang SW, Kim J, Yun JY, Jeong SH: High-performance NO2 gas sensor based on ZnO nanorod grown by ultrasonic irradiation. Sens Actuators B 2009, 141: 239–243. 10.1016/j.snb.2009.06.031View ArticleGoogle Scholar
- Ahn MW, Park KS, Hoe JH, Park JG, Kim DW, Choi KJ, Lee JH, Hong SH: Gas sensing properties of defect-controlled ZnO-nanowire gas sensor. Appl Phys Lett 2008, 93: 263103. 10.1063/1.3046726View ArticleGoogle Scholar
- Sadek AZ, Choopun S, Wlodarski W, Ippolito SJ, Kalantar-zadeh K: Characterization of ZnO nanobelt-based gas sensor for H2, NO2, and hydrocarbon sensing. IEEE Sens J 2007, 7: 919–924.View ArticleGoogle Scholar
- Baratto C, Sberveglieri G, Onischuk A, Caruso B, Stasio SD: Low temperature selective NO2 sensors by nanostructured fibres of ZnO. Sens Actuators B 2004, 100: 261–265. 10.1016/j.snb.2003.12.045View ArticleGoogle Scholar
- Ahn S, Park S, Ko H, Lee C: Enhanced NO2 gas sensing properties of WO3 nanorods encapsulated with ZnO. Appl Phys A 2012, 108: 53–58. 10.1007/s00339-012-7000-9View ArticleGoogle Scholar
- Landau O, Rothschild A, Zussman E: Processing-microstructure-properties correlation of ultrasensitive gas sensors produced by electrospinning. Chem Mater 2009, 21: 9–11. 10.1021/cm802498cView ArticleGoogle Scholar
- Kaur J, Kumar R, Bhatnagar MC: Effect of indium-doped SnO2 nanoparticles on NO2 gas sensing properties. Sens Actuators B 2007, 126: 478–484. 10.1016/j.snb.2007.03.033View ArticleGoogle Scholar
- Law M, Kind H, Messer B, Kim F, Yang P: Photochemical sensing of NO2 with SnO2 nanoribbon nanosensors at room temperature. Angew Chem Int Ed 2002, 41: 2405–2408. 10.1002/1521-3773(20020703)41:13<2405::AID-ANIE2405>3.0.CO;2-3View ArticleGoogle Scholar
- Zhang J, Wang S, Wang Y, Wang Y, Zhu B, Xia H, Guo X, Zhang S, Huang W, Wu S: NO2 sensing performance of SnO2 hollow-sphere sensor. Sens Actuators B 2009, 135: 610–617. 10.1016/j.snb.2008.09.026View ArticleGoogle Scholar
- Ramgir NS, Mulla IS, Vijayamohanan KP: A room temperature nitric oxide sensor actualized from Ru-doped SnO2 nanowires. Sens Actuators B 2005, 107: 708–715. 10.1016/j.snb.2004.12.073View ArticleGoogle Scholar
- Kaur J, Roy SC, Bhatnagar MC: Highly sensitive SnO2 thin film NO2 gas sensor operating at low temperature. Sens Actuators B 2007, 123: 1090–1095. 10.1016/j.snb.2006.11.031View ArticleGoogle Scholar
- Vomiero A, Bianchi S, Comini E, Faglia G, Ferroni M, Sberveglieri G: Controlled growth and sensing properties of In2O3 nanowires. Cryst Growth Des 2007, 7: 2500–2504. 10.1021/cg070209pView ArticleGoogle Scholar
- Xu P, Cheng Z, Pan Q, Xu J, Xiang Q, Yu W, Chu Y: High aspect ratio In2O3 nanowires: synthesis, mechanism and NO2 gas-sensing properties. Sens Actuators B 2008, 130: 802–808. 10.1016/j.snb.2007.10.044View ArticleGoogle Scholar
- Liu Z, Miyauchi M, Yamazaki T, Shen Y: Facile synthesis and NO2 gas sensing of tungsten oxdie nanorods assembled microspheres. Sens Actuators B 2009, 140: 514–519. 10.1016/j.snb.2009.04.059View ArticleGoogle Scholar
- Xia H, Wang Y, Kong F, Wang S, Zhu B, Guo X, Zhang J, Wu S: Au-doped WO3-based sensor for NO2 detection at low operating temperature. Sens Actuators B 2008, 134: 133–139. 10.1016/j.snb.2008.04.018View ArticleGoogle Scholar
- Teoh LG, Hon YM, Shieh J, Lai WH, Hon MH: Sensitivity properties of a novel NO2 gas sensor based on mesoporous WO3 thin film. Sens Actuators B 2003, 96: 219–225. 10.1016/S0925-4005(03)00528-8View ArticleGoogle Scholar
- Rahmani MB, Keshmiri S, Yu J, Sadek A, Al-Moafi A, Latham K, Li Y, Wlodarsk W, Kalantar-Zadeh K: Gas sensing properties of thermally evaporated lamellar MoO3. Sens Actuators B 2010, 145: 13–19. 10.1016/j.snb.2009.11.007View ArticleGoogle Scholar
- Guo Z, Li M, Liu J: Highly porous CdO nanowires: preparation based on hydroxy-and carbonate-containing cadmium compound precursor nanowires, gas sensing and optical properties. Nanotechnology 2008, 19: 245611. 10.1088/0957-4484/19/24/245611View ArticleGoogle Scholar
- Choi SW, Park JY, Kim SS: Synthesis of SnO2-ZnO core-shell nanofibers via a novel two-step process and their gas sensing properties. Nanotechnology 2009, 20: 465603–465608. 10.1088/0957-4484/20/46/465603View ArticleGoogle Scholar
- Chen IC, Lin SS, Lin TJ, Hsu CL, Hsueh TJ, Shieh TY: The assessment for sensitivity of a NO2 gas sensor with ZnGa2O4/ZnO core-shell nanowires-a novel approach. Sensors 2010, 10: 3057–3072. 10.3390/s100403057View ArticleGoogle Scholar
- Xu C, Tamaki J, Miura N, Yamazoe N: Grain size effect on gas sensitivity of porous SnO2-based elements. Sens Actuators B 1991, 3: 147–155. 10.1016/0925-4005(91)80207-ZView ArticleGoogle Scholar
- Park S, Ko H, Kim S, Lee C: Role of the interfaces in multiple networked one-dimensional core–shell nanostructured gas sensors. ACS Appl Mater Interfaces 2014, 6: 9595–9600. 10.1021/am501975vView ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited.