Biosensor for human IgE detection using shear-mode FBAR devices
© Chen et al.; licensee Springer. 2015
Received: 30 June 2014
Accepted: 6 January 2015
Published: 18 February 2015
Film bulk acoustic resonators (FBARs) have been evaluated for use as biosensors because of their high sensitivity and small size. This study fabricated a novel human IgE biosensor using shear-mode FBAR devices with c-axis 23°-tilted AlN thin films. Off-axis radio frequency (RF) magnetron sputtering method was used for deposition of c-axis 23°-tilted AlN thin films. The deposition parameters were adopted as working pressure of 5 mTorr, substrate temperature of 300°C, sputtering power of 250 W, and 50 mm distance between off-axis and on-axis. The characteristics of the AlN thin films were investigated by X-ray diffraction and scanning electron microscopy. The frequency response was measured with an HP8720 network analyzer with a CASCADE probe station. The X-ray diffraction revealed (002) preferred wurtzite structure, and the cross-sectional image showed columnar structure with 23°-tilted AlN thin films. In the biosensor, an Au/Cr layer in the FBAR backside cavity was used as the detection layer and the Au surface was modified using self-assembly monolayers (SAMs) method. Then, the antigen and antibody were coated on biosensor through their high specificity property. Finally, the shear-mode FBAR device with k t 2 of 3.18% was obtained, and the average sensitivity for human IgE detection of about 1.425 × 105 cm2/g was achieved.
KeywordsAlN Shear mode FBAR Biosensor
In recent years, piezoelectric materials have been used in surface acoustic wave (SAW) resonators [1-5] and film bulk acoustic wave resonators (FBARs) [6-10] because of their low cost, low weight, and good reproducibility. However, the SAW resonator has high insertion loss and poor power handling capability. Hence, this study evaluated the potential applications of FBARs for biosensors because of their advantages, including low insertion loss, good power handling, and small size. The FBAR devices were constructed by a piezoelectric layer sandwiched between two electrodes and attached to substrate with backside cavity. Piezoelectric materials such as zinc oxide (ZnO) and aluminum nitride (AlN) have been used in FBAR devices for various applications [11-13] owing to their high acoustic velocity, better quality factor, and high electromechanical coupling coefficient. Besides, the piezoelectric materials of ZnO and AlN can be combined with silicon technologies in semiconductor fabrication processes [14,15]. Moreover, the acoustic velocity of AlN is 10,400 m/s, and it suits application for FBAR devices.
The acoustic wave of a FBAR has two transmittance modes: longitudinal mode and shear mode. In shear mode, acoustic wave energy does not dissipate in a liquid environment . The backside cavity of FBAR can be used as the detection area for adsorbent matter. Under a mass loading, a frequency shift would be resulted in the frequency response of a FBAR . The analysis methods were used for biosensor in liquid and tiny mass detection in air through the shear mode and longitudinal mode, respectively. Thus, FBAR devices were fabricated and constructed to evaluate their potential use in biosensors.
Fabrication of FBAR devices
The characteristics of AlN thin films, including crystalline properties, preferred orientation, and cross-sectional morphologies were examined. The crystalline properties and preferred orientation of the AlN thin films were determined by X-ray diffraction scanning between 20° and 60° using a Siemens D5000 (Munich, Germany) with CuKα radiation. The surface morphologies and cross sections of AlN thin films were observed by field-emission scanning electron microscope (FESEM, JEOL-6700; JEOL Ltd., Akishima-shi, Japan). Finally, the frequency responses of FBAR devices with the biosensors were measured by HP8720 network analyzer.
FBAR devices for biosensor applications
Step (1): Use oxygen plasma process for Au surface cleaning.
Step (2): Inject cysteine solution (R.T., 1 h).
Step (3): Inject deionized (DI) water and dry using N2 gas.
Step (4): Inject glutaraldehyde solution (2.5%, R. T., 1 h).
Step (5): Inject DI water and dry with N2 gas.
Step (1): Wash with 200 μl phosphate-buffered saline (PBS) solution three times.
Step (2): Dip 200 μl diluted mouse anti-human IgE antibody (37°C, 2 h).
Step (3): Inject 200 μl, Tween-20 wash buffer three times.
Step (4): Inject 200 μl, 10 wt.% bovine serum albumin (BSA) solution (37°C, 0.5 h).
Step (5): Inject 200 μl, Tween-20 wash buffer three times.
Step (6): Inject 200 μl, diluted human IgE antigen with 0.707 μg/ml concentration.
where δm is the loading mass (9.1875 ng/cm2) and δf is the variation of the resonate frequency. Finally, the sensitivities of FBAR devices for human IgE detection were investigated.
Results and discussion
Structural and morphological properties of AlN thin films
Frequency responses of shear-mode FBAR devices
where V L and V S are the acoustic velocity, C 33 and C 44 are an elastic constant, and ρ is density of the wurtzite AlN. In this study, the practical acoustic velocity of longitudinal mode is 1.76 times than that of the shear mode, which is still consistent with the literature [25,27]. The electromechanical coupling coefficient (k t 2) of shear mode is a numerical measurement of the conversion efficiency between electrical and acoustic energy in piezoelectric materials. The k t 2 of the shear mode of the FBAR was calculated to be about 3.18%.
Frequency responses of biosensors for human IgE detection
The Au/Cr thin films were adopted as detection layer using a dual-gun DC sputtering system, the oxygen plasma process was used to clean the surface of the Au layer in order to improve the hydrophilic properties of the contact area between the bio-drop and Au layer [28-32].
The frequency shift and sensitivity of biosensors
Frequency shift, δf (MHz)
Sensitivity, S m (cm2/g)
1.41 × 105
1.44 × 105
The results of this study demonstrate that the proposed shear-mode FBAR device is highly promising for use in human IgE detection because of its high sensitivity, small size, low-cost, and rapid reaction process than conventional quartz crystal micro-balance (QCM) [37-41].
This study fabricated shear-mode FBAR devices for biosensor applications. The AlN thin films and Pt/Ti were adopted as the piezoelectric and electrode layers, respectively, in FBAR devices. The AlN thin films were fabricated at a working pressure of 5 mTorr, substrate temperature of 300°C, sputtering power of 250 W, and off-axis of 50 mm. The resulted AlN thin films exhibited a strong c-axis orientation and 23°-tilted. The obtained shear-mode FBAR devices had a frequency response of 1.175 GHz and a k t 2 of about 3.18%. For biosensor applications, the Au/Cr thin films were deposited on backside cavity of FBAR as bio-detection layer. The SAMs method was used for surface modification of Au thin films. Human IgE was detected by using a coating process to detect antibody with antigen. The average sensitivity for the shear-mode FBAR devices for human IgE detection was about 1.425 × 105 cm2/g.
The authors gratefully acknowledge the financial support from the National Science Council, the Republic of China (NSC Grant Numbers: No. NSC 102-2221-E-366-002, and NSC102-2221-E-110-029) and from the National Sun Yat-sen University (The Aim for the Top University Project, NSYSU).
- Jung JP, Lee JB, Kim JS, Park JS. Fabrication and characterization of high frequency SAW device with IDT/ZnO/AlN/Si configuration: role of AlN buffer. Thin Solid Films. 2004;447–448:605–9.View ArticleGoogle Scholar
- Legrani O, Elmazria O, Zhgoon S, Pigeat P, Bartasyte A. Packageless AlN/ZnO/Si structure for SAW devices applications. IEEE Sens J. 2013;13:487–91.View ArticleGoogle Scholar
- Meng X, Yang C, Chen Q, Gao Y, Yang J. Preparation of highly c-axis oriented AlN films on Si substrate with ZnO buffer layer by the DC magnetron sputtering. Mater Lett. 2013;90:49–52.View ArticleGoogle Scholar
- Kao KS, Cheng CC, Chung CJ, Chen YC. Surface acoustic wave properties of proton-exchanged LiNbO3 waveguides with SiO2 film. IEEE Trans Ultrason Ferroelectr Freq Control. 2005;52:503–6.View ArticleGoogle Scholar
- Wei CL, Chen YC, Cheng CC, Kao KS, Cheng DL, Cheng PS. Highly sensitive ultraviolet detector using a ZnO/Si layered SAW oscillator. Thin Solid Films. 2010;518:3059–62.View ArticleGoogle Scholar
- Clement M, Olivares J, Iborra E, González-Castilla S, Rimmer N, Rastogi A. AlN films sputtered on iridium electrodes for bulk acoustic wave resonators. Thin Solid Films. 2009;517:4673–8.View ArticleGoogle Scholar
- Lee JB, Jung JP, Lee MH, Park JS. Effects of bottom electrodes on the orientation of AlN films and the frequency responses of resonators in AlN-based FBARs. Thin Solid Films. 2004;447–448:610–4.View ArticleGoogle Scholar
- Yim M, Kim DH, Chai D, Yoon G. Effects of thermal annealing of W/SiO2 multilayer Bragg reflectors on resonance characteristics of film bulk acoustic resonator devices with cobalt electrodes. J Vac Sci Technol A. 2004;22:465–71.View ArticleGoogle Scholar
- Kirby PB, Potter MDG, Williams CP, Lim MY. Thin film piezoelectric property considerations for surface acoustic wave and thin film bulk acoustic resonators. J Eur Ceram Soc. 2003;23:2689–92.View ArticleGoogle Scholar
- Huang CL, Tay KW, Wu L. Fabrication and performance analysis of film bulk acoustic wave resonators. Mater Lett. 2005;59:1012–6.View ArticleGoogle Scholar
- Umar A, Rahman MM, Vaseem M, Hahn YB. Ultra-sensitive cholesterol biosensor based on low-temperature grown ZnO nanoparticles. Electrochem Commun. 2009;11:118–21.View ArticleGoogle Scholar
- Hong S, Yeo J, Manorotkul W, Kim G, Kwon J, An K, et al. Low-temperature rapid fabrication of ZnO nanowire UV sensor array by laser-induced local hydrothermal growth. J Nanomater. 2013;2013:246328.Google Scholar
- Akiyama M, Morofuji Y, Kamohara T, Nishikubo K, Tsubai M, Fukuda O, et al. Flexible piezoelectric pressure sensors using oriented aluminum nitride thin films prepared on polyethylene terephthalate films. J Appl Phys. 2006;100:114318.View ArticleGoogle Scholar
- Iborra E, Clement M, Capilla J, Olivares J, Felmetsger V. Low-thickness high-quality aluminum nitride films for super high frequency solidly mounted resonators. J Appl Phys. 2012;520:3060–3.Google Scholar
- Lin RC, Chen YC, Chang WT, Cheng CC, Kao KS. Highly sensitive mass sensor using film bulk acoustic resonator. Sensor Actuat A-Phys. 2008;147:425–9.View ArticleGoogle Scholar
- Zhang H, Kim ES. Micromachined acoustic resonant mass sensor. J Microelectromech S. 2005;14:699–706.View ArticleGoogle Scholar
- Chung CJ, Chen YC, Cheng CC, Kao KS. Synthesis and bulk acoustic wave properties on the dual mode frequency shift of solidly mounted resonators. IEEE Trans Ultrason Ferroelectr Freq Control. 2008;55:857–64.View ArticleGoogle Scholar
- Schwindt CD, Settipane R. Allergic rhinitis (AR) is now estimated to affect some 1.4 billion people globally and continues to be on the rise. Am J Rhinol. Allergy. 2012;26:S1–S1(1).Google Scholar
- Rai M, Carpinella MC. Naturally occurring bioactive compounds. Baker & Taylor Books: Elsevier Science; 2006. pp. 271.Google Scholar
- Chen YC, Chang WT, Cheng CC, Shen JY, Kao KS. Development of human IgE biosensor using Sezawa-mode SAW devices. Curr Appl Phys. 2014;14:608–13.View ArticleGoogle Scholar
- Huang IY, Lee MC. Development of a FPW allergy biosensor for human IgE detection by MEMS and cystamine-based SAM technologies. Sensor Actuat B-Chem. 2008;132:340–8.View ArticleGoogle Scholar
- Huang IY, Lee MC, Hsu CH, Wang CC. Development of a flexural plate-wave (FPW) immunoglobulin-E (IgE) allergy bio-sensing microsystem. Sensor Actuat B-Chem. 2012;162:184–93.View ArticleGoogle Scholar
- Huang IY, Lee MC, Chang YW. Development of a novel flexural plate wave biosensor for immunoglobulin-E detection by using SAM and MEMS technologies, 5th IEEE Conference on Sensors, October 2006. Daegu: IEEE; 2006. p. 70.Google Scholar
- Huang IY, Lee MC, Chang YW, Huang RS. Development and characterization of FPW based allergy biosensor, ISIE 2007. IEEE International Symposium on Industrial Electronics, June 2007. Daegu: IEEE; 2007. p. 2736.Google Scholar
- Qin L, Chen Q, Cheng H, Wang QM. Analytical study of dual-mode thin film bulk acoustic resonators (FBARs) based on ZnO and AlN films with tilted c-axis orientation. IEEE T Ultrason Ferr. 2010;57:840–1853.Google Scholar
- Chen YC, Chang WT, Kao KS, Yang CH, Cheng CC. The liquid sensor using thin film bulk acoustic resonator with c-axis tilted AlN films. J Nanomater. 2013;2013:245095.Google Scholar
- McNeil L, Grimsditch M, French RH. Vibrational spectroscopy of aluminum nitride. J Am Ceram Soc. 1993;76:1132–6.View ArticleGoogle Scholar
- Ostuni E, Chapman RG, Holmlin RE, Takayama S, Whitesides GM. A survey of structure–property relationships of surfaces that resist the adsorption of protein. Langmuir. 2001;17:5605–20.View ArticleGoogle Scholar
- Hook F, Rodahl M, Kasemo B, Brzezinski P. Structural changes in hemoglobin during adsorption to solid surfaces: effects of pH, ionic strength, and ligand binding. Proc Natl Acad Sci U S A. 1998;95:12271–6.View ArticleGoogle Scholar
- Kaufman ED, Belyea J, Johnson MC, Nicholson ZM, Ricks JL, Shah PK, et al. Probing protein adsorption onto mercaptoundecanoic acid stabilized gold nanoparticles and surfaces by quartz crystal microbalance and zeta-potential measurements. Langmuir. 2007;23:6053–62.View ArticleGoogle Scholar
- Tang Q, Xu CH, Shi SQ, Zhou LM. Formation and characterization of protein patterns on the surfaces with different properties. Synth Met. 2004;147:247–52.View ArticleGoogle Scholar
- Wang Y, Zhang Z, Jain V, Yi J, Mueller S, Sokolov J, et al. Potentiometric sensors based on surface molecular imprinting: detection of cancer biomarkers and viruses. Sensor Actuat B-Chem. 2010;146:381–7.View ArticleGoogle Scholar
- Chung CJ, Chen YC, Cheng CC, Wang CM, Kao KS. Superior dual mode resonances for 1/4 λ solidly mounted resonators, 2008 IEEE International Frequency Control Symposium, May 2008. Honolulu: IEEE; 2008. p. 250.Google Scholar
- Beghi MG. Acoustic waves-from microdevices to helioseismology. INTECH. 2011;14:501.Google Scholar
- Link M, Schreiter M, Weber J, Primig R, Pitzer D, Gabl R. Solidly mounted ZnO shear mode film bulk acoustic resonators for sensing applications in liquids. IEEE Trans Ultrason Ferroelectr Freq Control. 2006;53:492–6.View ArticleGoogle Scholar
- Yan Z, Zhou XY, Pang GKH, Zhang T, Liu WL, Cheng JG, et al. ZnO-based film bulk acoustic resonator for high sensitivity biosensor applications. Appl Phys Lett. 2007;90:143503-1–3.Google Scholar
- Weber J, Albers WM, Jussipekka T, Mathias L, Reinhard G, Wersing W, et al. Shear mode FBARs as highly sensitive liquid biosensors. Sensor Actuat A-Phys. 2006;128:84–8.View ArticleGoogle Scholar
- Wingqvist G, Bjurström J, Liljeholm L, Yantchev V, Katardjiev I. Shear mode AlN thin film electro-acoustic resonant sensor operation in viscous media. Sensor Actuat B-Chem. 2007;123:466–73.View ArticleGoogle Scholar
- Mathias L, Weber J, Schreiter M, Wersing W, Elmazria O, Alnot P. Sensing characteristics of high-frequency shear mode resonators in glycerol solutions. Sensor Actuat B-Chem. 2007;121:372–8.View ArticleGoogle Scholar
- Wingqvist G, Anderson H, Lennartsson C, Weissbach T, Yantchev V, Spetz AL. On the applicability of high frequency acoustic shear mode biosensing in view of thickness limitations set by the film resonance. Biosens Bioelectron. 2009;24:3387–90.View ArticleGoogle Scholar
- Fua YQ, Luo JK, Du XY, Flewitt AJ, Li Y, Markx GH, et al. Recent developments on ZnO films for acoustic wave based bio-sensing and microfluidic applications: a review. Sensor Actuat B-Chem. 2010;143:606–9.View ArticleGoogle Scholar
This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited.