Ultra-Sensitivity Glucose Sensor Based on Field Emitters
© to the authors 2009
Received: 4 May 2009
Accepted: 2 June 2009
Published: 14 June 2009
A new glucose sensor based on field emitter of ZnO nanorod arrays (ZNA) was fabricated. This new type of ZNA field emitter-based sensor shows high sensitivity with experimental limit of detection of 1 nM glucose solution and a detection range from 1 nM to 50 μM in air at room temperature, which is lower than that of glucose sensors based on surface plasmon resonance spectroscopy, fluorescence signal transmission, and electrochemical signal transduction. The new glucose sensor provides a key technique for promising consuming application in biological system for detecting low levels of glucose on single cells or bacterial cultures.
The glucose sensors are becoming an increasingly active area of research due to their applications in biological, chemical analyze, and clinical detection . Many glucose sensors based on surface plasmon resonance (SPR) spectroscopy [2, 3], fluorescence signal transmission (FST) [4–8], and electrochemical signal transduction (EST) [9–11], have been reported. However, the sensitivity of these glucose sensors was mostly confined to the millimolar, which restricted their application in the case of lower glucose concentration, such as cellular signal transduction and protein biosynthesis in single cell or bacterial cultures. In order to further promote the sensitivity, intensive efforts have been made in the exploration of a glucose sensors based on nanostructured materials such as carbon nanotubes, semiconductor quantum dots, and nanowires [12–16].
Due to their unique properties of good biological compatibility and stability, nanomaterials of metal oxides could play an important role in adsorption of biomolecules. They are able to couple with different redox enzymes, which make them particular promising for their applications on biosensors and bioelectronics [17, 18]. Being an important n-type semiconductor with a wide band gap (3.37 eV), ZnO possesses many versatile properties: high optical transparency, semiconducting, piezoelectric, nontoxicity, chemical stability, and electrochemical activity. Recently, one dimensional (1D) ZnO nanostructures for nanodevices such as field emitter , nanopiezotronics , gas and pH sensor , biosensor , transistor , and temperature laser , have received more and more attention due to their distinguished performance, high specific surface area, and facile preparation. 1D ZnO nanostructures have been used as electron mediators and adsorption matrices in amperometric biosensors . Sun et al. [22, 25, 26] constructed a glucose amperometric biosensor using 1D ZnO nanostructures as supporting materials for glucose oxidase (GO x ) loading, whose detection limit measured was 0.02 and 0.01 mM. Ren et al.  fabricated a ZnO nanorod-gated AlGa/GaN high electron mobility transistor for the detection of glucose, which showed a linear range from 0.01 to 3.45 mM. In this contribution, we present a new type of nanodevice, a glucose sensor based on field emitter of ZnO nanorod arrays (ZNA). The obvious changes of field emission properties of ZNA caused by surface energy band bending, which induced by surface adsorptions. This new ZNA field emitter-based sensor shows high sensitivity with an experimental limit of detection (LOD) of 1 nM glucose solution and a wide detection range from 1 nM to 50 μM in air at room temperature, which is lower than that of glucose sensor based on FST and EST.
Materials and Apparatus
GO x (100 U mg−1) was purchased from Amersco Inc. (USA). All other reagents (analytical-grade) were purchased from Beijing Chemical Regent Company. The buffer solution was phosphate buffer (6 mM, pH 7.4). The water was purified using a Millipore filtration system.
The arrays of ZNA are characterized by field emission scanning electron microscopy (FESEM, Hitachi, S-4300). The field emission properties of ZNA are measured using a two-parallel-plate configuration in a homemade vacuum chamber at a base pressure of ~1.0×10−6 Pa at room temperature. The sample is attached to one of stainless-steel plates as cathode with the other plate as anode. The distance between the electrodes is 300 μm. A direct current voltage sweeping from 0 to 5,000 V was applied to the sample at a step of 50 V. The emission current is monitored using a Keithley 6485 picoammeter.
Synthesis of ZNA and Glucose Sensors
ZNA were directly grown on a Si (100) wafer with an area of 4 cm2 by a multi-step hydrothermal process . The Si (100) wafer (4 cm2) is equably cut as 16 pieces of same area (0.5×0.5 cm2) samples. The samples for measurement of field emission were then prepared as follow: the films of Si (100) wafers deposited with ZNA were immersed in 2 mL Eppendorf cups with 10 μL of glucose oxidase (GO x ) (10 U μL−1) and 1 mL of PBS buffer solution, respectively. Then the β-d-(+) glucose with different concentrations (from 1 nM to 50 μM) was added into the solution and incubated for 30 min at room temperature. After incubation, the samples were washed by pure water for three times. The samples were desiccated via vacuum for field emission measurement.
Results and Discussion
The properties of ZnO nanostructures are significantly influenced by surface adsorptions, which have been attracting great attention because the surface absorptions sometime disturb the fluorescence, field emission, and field effect transistors [30–32]. In general, the changes of properties caused by surface energy band bending, induced by surface adsorptions. As shown in Fig. 4, GO x specifically catalyzes the oxidation of β-d-(+)-glucose into gluconate and H2O2. H2O2 and ZnO to form ZnO(OH) x on the surface of ZnO nanorods , which depletes the electrons on the ZnO nanorods and yields oxygen ions (O−, O2−, or O2−) . Leading to the electrons on the ZnO nanorods are trapped by the adsorbed oxygen molecules, and the surface depletion region of ZnO nanorods can be formed, making resistance increase. While the H2O2 is raised with the increase of glucose, more electrons are captured by the oxygen molecules at the nanorod surface. Thus, the surface depletion region is widened and the carrier density in the ZnO nanorod is decreased even more. At the same time, gluconolactone can be absorbed strongly on the surface of ZnO nanorods by hydrogen bonding with ZnO(OH) x , which induces the surface passivation to block the electrons emission of ZnO nanorods under electronic field. The surface passivation increases to completely deplete the electrons emission of ZnO nanorods with the increase of ZnO(OH) x on the surface of ZnO nanorods. The result indicates that field emission properties were not observed on the ZNA at the glucose concentration ≥500 nM.
In summary, we demonstrated a new glucose sensor based on the field emitter of ZNA. The results showed that a wide range of glucose concentrations from 1 nM to 50 μM is easily detected, which exhibits ultra-sensitivity for glucose detection. The experimental LOD of glucose concentration is 1 nM, which is lower than previous reported glucose sensor based on EST, FST, and SPR. The new glucose sensor shows the potential to detect low levels of glucose in biological system.
This work was supported by the National Nature Science Foundation of China (10874187 and 20873155) and the National Basic Research 973 Program of China.
- Rakow NA, Suslick KS: Nature. 2000, 406: 710. COI number [1:CAS:528:DC%2BD3cXmt1CgsLw%3D]; Bibcode number [2000Natur.406..710R] 10.1038/35021028View ArticleGoogle Scholar
- Lee B, Perez-Luna VH: Anal. Chem.. 2005, 77: 7204. COI number [1:CAS:528:DC%2BD2MXhtFSjtrnN] 10.1021/ac050484nView ArticleGoogle Scholar
- Aslan K, Lakowicz JR, Geddes CD: Anal. Chem.. 2005, 77: 2007. COI number [1:CAS:528:DC%2BD2MXhtFeht7c%3D] 10.1021/ac0484880View ArticleGoogle Scholar
- Ge X-D, Tolosa L, Rao G: Anal. Chem.. 2004, 76: 1403. COI number [1:CAS:528:DC%2BD2cXksFChtw%3D%3D] 10.1021/ac035063pView ArticleGoogle Scholar
- Meadows DL, Schultz JS: Talanta. 1988, 35: 145. COI number [1:CAS:528:DyaL1cXitFygtL8%3D] 10.1016/0039-9140(88)80053-5View ArticleGoogle Scholar
- Blagoi G, Rosenzweig N, Rosenzweig Z: Anal. Chem.. 2005, 77: 393. COI number [1:CAS:528:DC%2BD2cXhtVCmsLnO] 10.1021/ac048736sView ArticleGoogle Scholar
- He F, Tang Y, Yu M, Wang S, Li Y, Zhu D: Adv. Funct. Mater.. 2006, 16: 91. COI number [1:CAS:528:DC%2BD28XpvVWrsA%3D%3D] 10.1002/adfm.200500602View ArticleGoogle Scholar
- He F, Feng F, Wang S, Li Y, Zhu D: J. Mater. Chem.. 2007, 17: 3702. COI number [1:CAS:528:DC%2BD2sXps1Sns78%3D] 10.1039/b703856aView ArticleGoogle Scholar
- Chen W, Yao H, Tzang CH, Zhu J, Yang M, Lee S-T: Appl. Phys. Lett.. 2006, 88: 213104. Bibcode number [2006ApPhL..88u3104C] 10.1063/1.2206102View ArticleGoogle Scholar
- Chen T, Friedman KA, Lei I, Heller A: Anal. Chem.. 2000, 72: 3757. COI number [1:CAS:528:DC%2BD3cXks1ehtr0%3D] 10.1021/ac000348cView ArticleGoogle Scholar
- Sapre AG, Bedekar A, Deshpande AV, Lali AM: Biotechnol. Lett.. 2000, 22: 569. COI number [1:CAS:528:DC%2BD3cXjtlKmtLg%3D] 10.1023/A:1005624528652View ArticleGoogle Scholar
- Wang J, Musameh M, Lin YH: J. Am. Chem. Soc.. 2003, 125: 2408. COI number [1:CAS:528:DC%2BD3sXpvVartA%3D%3D] 10.1021/ja028951vView ArticleGoogle Scholar
- Duong HD, Il Rhee J: Talanta. 2007, 73: 899. COI number [1:CAS:528:DC%2BD2sXht1SrurjO] 10.1016/j.talanta.2007.05.011View ArticleGoogle Scholar
- Curreli M, Li C, Sun YH, Lei B, Gundersen MA, Thompson ME, Zhou CW: J. Am. Chem. Soc.. 2005, 127: 6922. COI number [1:CAS:528:DC%2BD2MXjs1Sitr8%3D] 10.1021/ja0503478View ArticleGoogle Scholar
- Liu Y, Wang M, Zhao F, Xu Z, Dong S: Biosens. Bioelectron.. 2005, 21: 984. COI number [1:CAS:528:DC%2BD2MXhtFKhsbjF] 10.1016/j.bios.2005.03.003View ArticleGoogle Scholar
- Yeh JI, Lazareck A, Ho Kim J, Xu J, Du S: Biosens. Bioelectron.. 2007, 23: 568. COI number [1:CAS:528:DC%2BD2sXhtF2qt77K] 10.1016/j.bios.2007.06.019View ArticleGoogle Scholar
- Topoglidis E, Cass AEG, ORegan B, Durrant JR: J. Electroanal. Chem.. 2001, 517: 20. COI number [1:CAS:528:DC%2BD3MXovFCmtLg%3D] 10.1016/S0022-0728(01)00673-8View ArticleGoogle Scholar
- Yang YH, Yang HF, Yang MH, Liu YL, Shen GL, Yu RQ: Anal. Chim. Acta.. 2004, 525: 213. COI number [1:CAS:528:DC%2BD2cXoslSms70%3D] 10.1016/j.aca.2004.07.071View ArticleGoogle Scholar
- Xu CX, Sun XW: Appl. Phys. Lett.. 2003, 83: 3806. COI number [1:CAS:528:DC%2BD3sXosF2ltrs%3D]; Bibcode number [2003ApPhL..83.3806X] 10.1063/1.1625774View ArticleGoogle Scholar
- Wang ZL, Song JH: Science. 2006, 312: 242. COI number [1:CAS:528:DC%2BD28XjtlKqu7g%3D]; Bibcode number [2006Sci...312..242W] 10.1126/science.1124005View ArticleGoogle Scholar
- Wang HT, Kang BS, Ren F, Tien LC, Sadik PW, Norton DP, Pearton SJ, Lin J: J. Appl. Phys. Lett.. 2005, 86: 243503. Bibcode number [2005ApPhL..86x3503W] 10.1063/1.1949707View ArticleGoogle Scholar
- Wang JX, Sun XW, Wei A, Lei Y, Cai XP, Li CM, Dong ZL: Appl. Phys. Lett.. 2006, 88: 233106. Bibcode number [2006ApPhL..88w3106W] 10.1063/1.2210078View ArticleGoogle Scholar
- Fan ZY, Lu JG: Appl. Phys. Lett.. 2005, 86: 123510. Bibcode number [2005ApPhL..86l3510F] 10.1063/1.1883715View ArticleGoogle Scholar
- Sun XW, Yu SF, Xu CX, Yuen C, Chen BJ, Li S: Jpn. J. Appl. Phys. 2.. 2003, 42: L1229. COI number [1:CAS:528:DC%2BD3sXosFSitbo%3D] 10.1143/JJAP.42.L1229View ArticleGoogle Scholar
- Wei A, Sun XW, Wang JX, Lei Y, Cai XP, Li CM, Dong ZL, Huang W: Appl. Phys. Lett.. 2006, 89: 123902. Bibcode number [2006ApPhL..89l3902W] 10.1063/1.2356307View ArticleGoogle Scholar
- Zang J, Li CM, Cui X, Wang JX, Sun XW, Dong H, Sun CQ: Electroanalysis. 2007, 19: 1008. COI number [1:CAS:528:DC%2BD2sXls1ymtr4%3D] 10.1002/elan.200603808View ArticleGoogle Scholar
- Kang BS, Wang HT, Rena F, Pearton SJ, Morey TE, Dennis DM, Johnson JW, Rajagopal P, Roberts JC, Piner EL, Linthicum KJ: Appl. Phys. Lett.. 2007, 91: 252103. Bibcode number [2007ApPhL..91y2103K] 10.1063/1.2825574View ArticleGoogle Scholar
- Greene LE, Law M, Goldberger J, Kim F, Johnson JC, Zhang YF, Saykally RJ, Yang PD: Angew. Chem. Int. Ed.. 2003, 42: 3031. COI number [1:CAS:528:DC%2BD3sXlslGksbw%3D] 10.1002/anie.200351461View ArticleGoogle Scholar
- Bao SJ, Li CM, Zang JF, Cui XQ, Qiao Y, Guo J: Adv. Funct. Mater.. 2008, 18: 591. COI number [1:CAS:528:DC%2BD1cXjsVKmu7o%3D] 10.1002/adfm.200700728View ArticleGoogle Scholar
- Bohle DS, Spina CJ: J. Am. Chem. Soc.. 2007, 129: 12380. COI number [1:CAS:528:DC%2BD2sXhtVKmsbnE] 10.1021/ja0747223View ArticleGoogle Scholar
- Zhang Y, Kolmakov A, Chretien S, Metiu H, Moskovits M: Nano Lett.. 2004, 4: 403. COI number [1:CAS:528:DC%2BD2cXosFKqsA%3D%3D]; Bibcode number [2004NanoL...4..403Z] 10.1021/nl034968fView ArticleGoogle Scholar
- Song S, Hong WK, Kwon SS, Lee T: Appl. Phys. Lett.. 2008, 92: 263109. Bibcode number [2008ApPhL..92z3109S] 10.1063/1.2955512View ArticleGoogle Scholar
- Li QH, Liang YX, Wan Q, Wang TH: Appl. Phys. Lett.. 2004, 85: 6389. COI number [1:CAS:528:DC%2BD2cXhtFSgu7rI]; Bibcode number [2004ApPhL..85.6389L] 10.1063/1.1840116View ArticleGoogle Scholar