- Nano Express
- Open Access
Salivary Electrochemical Cortisol Biosensor Based on Tin Disulfide Nanoflakes
Nanoscale Research Letters volume 14, Article number: 189 (2019)
Cortisol, a steroid hormone, is secreted by the hypothalamic-pituitary-adrenal system. It is a well-known biomarker of psychological stress and is hence known as the “stress hormone.” If cortisol overexpression is prolonged and repeated, dysfunction in the regulation of cortisol eventually occurs. Therefore, a rapid point-of-care assay to detect cortisol is needed. Salivary cortisol electrochemical analysis is a non-invasive method that is potentially useful in enabling rapid measurement of cortisol levels. In this study, multilayer films containing two-dimensional tin disulfide nanoflakes, cortisol antibody (C-Mab), and bovine serum albumin (BSA) were prepared on glassy carbon electrodes (GCE) as BSA/C-Mab/SnS2/GCE, and characterized using electrochemical impedance spectroscopy and cyclic voltammetry. Electrochemical responses of the biosensor as a function of cortisol concentrations were determined using cyclic voltammetry and differential pulse voltammetry. This cortisol biosensor exhibited a detection range from 100 pM to 100 μM, a detection limit of 100 pM, and a sensitivity of 0.0103 mA/Mcm2 (R2 = 0.9979). Finally, cortisol concentrations in authentic saliva samples obtained using the developed electrochemical system correlated well with results obtained using enzyme-linked immunosorbent assays. This biosensor was successfully prepared and used for the electrochemical detection of salivary cortisol over physiological ranges, based on the specificity of antibody-antigen interactions.
Cortisol, a steroid hormone, is secreted by the hypothalamic-pituitary-adrenal (HPA) system. It is a well-known biomarker of psychological stress and hence called the “stress hormone” [1, 2]. Cortisol levels follow a circadian rhythm over a 24-h cycle; the highest levels are observed early morning, and the levels progressively reduce by night [3,4,5,6]. Excessive levels of cortisol can cause Cushing’s disease, with symptoms of central obesity, purple striae, and proximal muscle weakness. However, reduced levels of cortisol can lead to Addison’s disease, with chronic fatigue, malaise, anorexia, postural hypotension, and hypoglycemia [7,8,9]. Therefore, maintaining appropriate cortisol balance is essential for human health.
A growing interest in the measurement of cortisol as a precursor to medically and psychologically relevant events has developed, among which the most recent affliction is post-traumatic stress disorder (PTSD). The importance of aberrant HPA axis function in PTSD is indisputable; hence, traditional assessment methods are still able to provide abundant evidence and information [10,11,12,13,14]. Recently, many studies have reported the importance of cortisol detection and have identified correlations with different illnesses [15,16,17,18]. Various studies have confirmed that cortisol is related to autism spectrum disorder , depression , suicidal ideation , childhood adversity, and externalizing disorders .
Although identifying cortisol levels represents an important diagnostic tool, routine laboratory cortisol detection techniques such as chromatography [23, 24], radioimmunoassay , electro-chemiluminescent immunoassay [26,27,28], enzyme-linked immunosorbent assay [28, 29], surface plasmon resonance [1, 30, 31], and quartz crystal microbalance  involve extensive analysis time, are expensive, and cannot be implemented in point-of-care (POC) settings . Therefore, there is currently a need for sensitive, efficient, and real-time determination of cortisol levels.
In recent years, electrochemical immunoassay methods, which are established on the specific molecular recognition between antigens and antibodies, have emerged as a promising technology due to salient characteristics, such as involving simple devices, rapid analysis, low cost, label-free POC testing, high sensitivity, and low detection thresholds for cortisol in bio-fluids [34, 35]. Electrical potential changes are ascribed to variations in the concentration of electrochemical redox reactions at the electrode. Secreted cortisol eventually enters the circulatory system and can be found in various bio-fluids such as interstitial fluid , blood , urine , sweat , and saliva . The advantages of electrochemical detection of salivary cortisol, which is a non-invasive method, with easy sample collection, handling, and storage, have enhanced its potential for application in POC sensors for real-time measurement .
An ideal biosensor should have low detection limits, rapid selectivity, and high sensitivity. In order to fabricate an immunosensor, the immobilizing matrix chosen should possess high surface functionality, high biomolecule loading, and low resistance to electron transport, with a high electron transfer rate . However, metal sulfide nanomaterials have been rarely suggested for the immobilization of proteins for electrochemical biosensing. Therefore, here, tin disulfide was selected as a potential immobilizing matrix for immunosensor development in order to detect cortisol present in saliva.
Nano two-dimensional (2D) materials have attracted abundant research interests in the recent decade. There are a variety of kinds of 2D materials ranging from semiconductor to metal and from inorganic to organic [43,44,45,46] and related composite [47,48,49,50]. The discovery, manufacturing, and investigation on nano 2D material are prevailing streams in various fields. Nano 2D tin disulfide (SnS2), an n-type semiconductor with a bandgap of 2.18–2.44 eV [51, 52], consists of Sn atoms sandwiched between two layers of hexagonally disposed and closely arranged sulfur (S) atoms, with adjacent S layers linked by weak van der Waals forces . Because of its intriguing electrical properties, high carrier mobility, good chemical stability, low cost, and optical properties , SnS2 has evolved into a promising material for various applications in solar cells and optoelectronic devices [55, 56], as electrodes in lithium-ion batteries [57, 58], gas sensors, and glucose monitors [59, 60]. The selection of electrode material is an important key factor to improve the performance by providing a large reaction area and favorable microenvironment for facilitating electron transfer between enzyme and electrode surface.
In this work, biosensors were fabricated using SnS2 as the immobilizing matrix to detect cortisol. The results of differential pulse voltammetry (DPV) studies related to electrochemical sensing show a high sensitivity of 0.0103 mA/Mcm2 and the lowest detection concentration of 100 pM.
Materials and Methods
Hydrocortisone (cortisol), anti-rabbit cortisol antibody (anti-cortisol, C-Mab), potassium hexacyanoferrate (II), potassium hexacyanoferrate (III), β-estradiol, testosterone, progesterone, and corticosterone were purchased from Sigma-Aldrich (St. Louis, MO, USA). Bovine serum albumin (BSA) was obtained from PanReac. Tin (IV) chloride pentahydrate (SnCl4.5H2O) and thioacetamide (C2H5NS) were supplied by Showa (Japan) and Alfa Aesar (UK). Phosphate buffered saline (PBS) prepared with NaCl, KCl, Na2HPO4, and KH2PO4 were purchased from Sigma-Aldrich. Micro-polished alumina was sourced from Buehler (UK). All other chemicals were of analytical grade and were used without further purification. Cortisol Saliva ELISA kit (Cat # SA E-6000) was purchased from LDN (Germany).
Synthesis of Tin Disulfide
Powders of SnCl4·5H2O and C2H5NS were mixed in 70 mL deionized water and adjusted pH to 7.4. A hydrothermal autoclave reactor containing the reactants was heated from room temperature to 200 °C in 1 h, and maintained at 200 °C for 11 h. Then, the resulting SnS2 powder was washed with deionized water and ethanol at 6000 rpm for 15 min, and finally dried in air at 80 °C. This hydrothermal method was successfully applied for the synthesis of SnS2.
X-ray diffraction (XRD, PANalytical, The Netherlands) was utilized to investigate the crystal phase of 2D hexagonal SnS2 flakes. Multi-functional field emission scanning electron microscopy (FE-SEM, Zeiss, Germany) was used to image the surface morphology of materials. Field emission gun transmission electron microscopy (FEG-TEM, Tecnai, USA) was used to discern the microstructure of SnS2, and selected area diffraction (SAED, Tecnai) was used to obtain crystal patterns.
Fabrication of BSA/C-Mab/SnS2/GCE Biosensors
Glassy carbon electrodes (GCEs) were first polished with alumina slurry, and then drops of a mixture of 5 M SnS2 were deposited on the surface of pretreated GCEs. Solutions of anti-cortisol antibody (1 mg/mL) and BSA (1%) were prepared in PBS. SnS2/GCE was then decorated with the antibody and BSA solutions in sequence. The fabricated BSA/C-Mab/SnS2/GCE biosensors were stored under refrigeration at 4 °C when not in use. The research concept and setup of detection system are illustrated in Fig. 1.
Fabricated BSA/C-Mab/SnS2/GCEs were characterized using electrochemical impedance spectroscopy (EIS) and cyclic voltammetry (CV) to compare their electro-active behaviors. Electrochemical response studies as a function of cortisol concentration were conducted using CV and differential pulse voltammetry (DPV). All the experiments were performed using a three-electrode system with a GCE as the working electrode, a Pt wire as the auxiliary electrode, and a saturated calomel electrode as the reference electrode in 10 mM PBS (pH 7.4) containing 5 mM Fe(CN)63-/4-. Electrochemical measurements were performed on a Model CHI6114E series electrochemical workstation (CH Instruments, USA). The CV and DPV measurements were carried out between − 0.4 V and 1.0 V at 10 mV/s scan rate, unless specified otherwise.
Saliva Sample Collection and Electrochemical Sensing
Saliva sample (2 mL) was collected from two healthy voluntary subjects at around noon for validating the developed BSA/C-Mab/SnS2/GCE. Saliva samples were obtained without any filtration and initially stored at − 20 °C for maintaining biological characteristics. Before sensing, the saliva samples were thawed to room temperature and centrifuged at 3500 rpm for 15 min to collect the supernatant for measurement. The separated saliva was stored at − 20 °C. The BSA/C-Mab/SnS2/GCE was utilized for the electrochemical sensing of cortisol concentrations in saliva samples. The detection of cortisol using electrochemical analysis with the BSA/C-Mab/SnS2/GCE was compared with that of the commercially available ELISA cortisol kit mentioned above.
The inhibitory effect of potential confounding agents, such as other steroid hormones, on BSA/C-Mab/SnS2/GCE specificity was investigated by placing the biosensor in the following different solutions: 100 nM β-estradiol, 100 nM testosterone, 100 nM progesterone, and 100 nM corticosterone, for 10 min and then scanned by CV. The scanning rate was 10 mV/s and the scanning range was from − 0.4 V to 0.6 V.
Detection of Salivary Cortisol by ELISA
ELISA was performed on the saliva samples according to the manufacturer’s protocol. To establish a calibration curve for cortisol measurements, the assay was performed in a 96-well titer plate containing six known standard cortisol concentrations (0.0, 0.1, 0.4, 1.7, 7.0, and 30 ng/mL) for determining the absorbance of each well at 450 nm. The calibration curve was fitted with a trendline to obtain an equation for the calculation of unknown samples.
Results and Discussion
Material Analysis of SnS2
As seen from the XRD pattern in Fig. 2a, the as-synthesized product displays only the XRD peaks corresponding to the hexagonal phase SnS2 (JCPDS card no. 89-2358). Figure 2b, c illustrates the FE-SEM images of the as-synthesized SnS2 having uniform flake-like morphology with a size of approximately 300 nm. Figure 2d–f shows the FEG-TEM and SAED images of SnS2, in which lattice fringe spacings of 0.167 nm and 0.316 nm are identified for hexagonal SnS2 as a single crystalline structure. The stacking of nanoflakes is less than 10 layers with a total thickness of less than 10 nm.
Electrochemical Responses of the Electrode
Oxidation current can greatly increase by the addition of tin disulfide. As shown in Fig. 3a, b, the magnitude of the oxidation current reduced from SnS2/GCE to C-Mab/SnS2/GCE, followed by BSA/C-Mab/SnS2/GCE, as the charge transfer resistance value increased. Therefore, the results indicate that the sensor properties were modified on the electrode. Initially, BSA/C-Mab/SnS2/GCE was studied by varying the scan rate from 10 mV/s to 100 mV/s, as shown in Fig. 3c. The change in current response with scan rate, as plotted in Fig. 3d, shows that the oxidation current increased linearly with scan rate, and followed the relation: I = 0.5156 υ–0.0319 (R2 = 0.9985) in oxidation, and I = 0.6758υ–0.0288 (R2 = 0.9997) in reduction. However, near-linearity for the increment in peak current with increasing scan rate with well-defined redox peaks indicates a surface-controlled process, with stable electron transfer.
The current decreased with increasing concentration of cortisol over the range of 100 pM to 100 μM. The difference in current directly correlated to the cortisol concentration being sensed. Current values and well-separated oxidation peaks were obtained for BSA/C-Mab/SnS2/GCE electrodes, as shown in Fig. 3e, f. The change in current with the log of concentration was nearly linear. It is clear that the reduction in the linear regression coefficient is better for CV. Therefore, further measurements were made with more specific and accurate DPV. The results of such DPV studies indicated that the magnitude of current response decreased with the addition of cortisol, as illustrated in Fig. 3g. A calibration curve presented in Fig. 3h plots the magnitude of current response and logarithm of cortisol concentration, and was found to be linearly dependent and to follow the equation: y = − 0.0103x + 0.0443; R2 = 0.9979. This sensor exhibited a detection range between 100 pM to 100 μM, with a limit of detection of 100 pM and a sensitivity of 0.0103 mA/Mcm2 (R2 = 0.9979).
Storage Stability Study
CV studies were also carried out to study the shelf life of the BSA/C-Mab/SnS2/GCE at intervals of 1 day to 1 week. In order to compare two preservation conditions, one condition was to store the electrodes dried under vacuum, while the other was to store the electrodes at 4 °C. The redox peak stability of the electrodes at 4 °C and under vacuum are shown in Fig. 4a, c, respectively. It is clear that the preservation condition at 4 °C was better than that under vacuum. Figure 4b, d shows that the electrode stability value was 82% with the electrodes stored under vacuum for 7 days, while the electrode stability value was 91% with the electrodes stored at 4 °C. It can be observed that the stability of electrodes stored at 4 °C was higher than that under vacuum. The loss of activity of the electrode was possibly caused by degradation of the cortisol antibody activity under vacuum. The storage stability is a crucial issue for enzymatic sensor. A protective coating may be introduced in the future design of the electrode.
The results of CV studies of BSA/C-Mab/SnS2/GCE for measuring potential confounding agents, such as β-estradiol (100 nM), testosterone (100 nM), progesterone (100 nM), and corticosterone (100 nM) with respect to cortisol (10 nM), are shown in Fig. 5a. Compared to the change in the response of the cortisol signal, the effects of interference were less than 5% of the result for cortisol, suggesting that such potential interferences can be conveniently neglected.
Detection of Salivary Cortisol Using ELISA and Electrochemical Methods
Measurements of salivary cortisol samples performed with ELISA and the BSA/C-Mab/SnS2/GCE electrode are summarized in Table 1 and Fig. 5b. The concentrations of cortisol determined using ELISA were 1.105 ×10−8 M and 3.998 × 10−9 M. The calculated results of cortisol using electrochemical measurement were 1.046 × 10−8 M and 3.911 × 10−9 M. Good correlation was achieved with these two techniques, exhibiting comparable results with only a 2–5% difference. Hence, the results demonstrate that this BSA/C-Mab/SnS2/GCE can be employed for electrochemical cortisol sensing in biologically relevant fluids such as saliva.
Comparison with Other Studies
The results of this study were compared with other studies involving electrochemical sensors of salivary cortisol reported in the literature in order to gain a better understanding of the performance of this BSA/C-Mab/SnS2/GCE. Tables 2 and 3 show comparisons of results obtained using non-gold electrodes in cortisol detection. There are three main advantages of the present work. First, the materials are much lower in cost than the devices presented in other studies. Second, the preparation process was relatively simple and rapid. Finally, the detection limit was similar to that reported in other literature or, indeed, even better than those reported, while the target detection range for salivary cortisol is easily obtained.
A hydrothermal method has been successfully applied for the synthesis of SnS2. The properties of SnS2 were characterized by XRD, FE-SEM, FEG-TEM, and SAED. Electrochemical responses of the electrode as a function of cortisol concentrations were determined using CV and DPV. Our cortisol sensor exhibited a detection range from 100 pM to 100 μM, a detection limit of 100 pM, and sensitivity of 0.0103 mA/Mcm2 (R2 = 0.9979). The obtained sensing parameters were in normal physiological ranges. The impact of potential interference was less than 5%, indicating good specificity of this sensor. Stability testing demonstrated that the activity of the sensor stored at 4 °C was better than under vacuum. The results of this electrode for the measurement of cortisol in saliva samples were consistent with ELISA. Therefore, electrochemical analysis using this BSA/C-Mab/SnS2/GCE electrode can replace more traditional time-consuming immunoassay approaches.
Availability of Data and Materials
All data generated or analyzed during this study are included in this published article.
Bovine serum albumin
- C-Mab :
Differential pulse voltammetry
Electrochemical impedance spectroscopy
Enzyme-linked immunosorbent assay
Field emission gun transmission electron microscope
Field emission scanning electron microscope
Glassy carbon electrodes
Phosphate buffered saline
Post-traumatic stress disorder
Selected area diffraction
Stevens RC, Soelberg SD, Near S, Furlong CE (2008) Detection of cortisol in saliva with a flow-filtered, portable surface plasmon resonance biosensor system. Anal Chem 80:6747–6751
de Kloet ER, Joels M, Holsboer F (2005) Stress and the brain: from adaptation to disease. Nat Rev Neurosci 6:463–475
Corbalán-Tutau D, Madrid JA, Nicolás F, Garaulet M (2014) Daily profile in two circadian markers “melatonin and cortisol” and associations with metabolic syndrome components. Physiol Behav 123:231–235
Nicolson NA (2008) Measurement of Cortisol. In: Luecken LJ, Gallo LC (eds) Handbook of Physiological Research Methods in Health Psychology. SAGE, Thousand Oaks
Ramsay D, Lewis M (2003) Reactivity and regulation in cortisol and behavioral responses to stress. Child Dev 74:456–464
Kaushik A, Vasudev A, Arya SK, Pasha SK, Bhansali S (2014) Recent advances in cortisol sensing technologies for point-of-care application. Biosens Bioelectron 53:499–512
Mazziotti G, Gazzaruso C, Giustina A (2011) Diabetes in Cushing syndrome: basic and clinical aspects. Trends Endocrinol Metab 22:499–506
Chakera AJ, Vaidya B (2010) Addison Disease in Adults: Diagnosis and Management. Am J Med 123:409–413
Gatti R, Antonelli G, Prearo M, Spinella P, Cappellin E, De Palo EF (2009) Cortisol assays and diagnostic laboratory procedures in human biological fluids. Clin Biochem 42:1205–1217
Meewisse ML (2007) Cortisol and post-traumatic stress disorder in adults: systematic review and metaanalysis. Br J Psychiatry 191:387–392
Lee J-H, Jung H-I (2013) Biochip technology for monitoring posttraumatic stress disorder (PTSD). BioChip J 7:195–200
Arya SK, Chornokur G, Venugopal M, Bhansali S (2010) Dithiobis(succinimidyl propionate) modified gold microarray electrode based electrochemical immunosensor for ultrasensitive detection of cortisol. Biosens Bioelectron 25:2296–2301
Morris MC, Hellman N, Abelson JL, Rao U (2016) Cortisol, heart rate, and blood pressure as early markers of PTSD risk: a systematic review and meta-analysis. Clin Psychol Rev 49:79–91
Stoppelbein L, Greening L, Fite P (2012) The role of cortisol in PTSD among women exposed to a trauma-related stressor. J Anxiety Disord 26:352–358
Landay A, Patterson S, Moran P, Epel E, Sinclair E, Kemeny ME, Deeks SG, Bacchetti P, Acree M, Epling L, Kirschbaum C, Hecht FM (2013) Cortisol Patterns Are Associated with T Cell Activation in HIV. PLoS ONE 8:e63429
Djamshidian A, O'Sullivan SS, Papadopoulos A, Bassett P, Shaw K, Averbeck BB, Lees A (2011) Salivary cortisol levels in Parkinson’s disease and its correlation to risk behaviour. J Neurol Neurosurg Psychiatry 82:1107–1111
Djamshidian A, Averbeck BB, Lees AJ, O'Sullivan SS (2011) Clinical aspects of impulsive compulsive behaviours in Parkinson's disease. J Neurol Sci 310:183–188
Singh A, Kaushik A, Kumar R, Nair M, Bhansali S (2014) Electrochemical sensing of cortisol: a recent update. Appl Biochem Biotechnol 174:1115–1126
Sharpley CF, Bitsika V, Andronicos NM, Agnew LL (2016) Is afternoon cortisol more reliable than waking cortisol in association studies of children with an ASD? Physiol Behav 155:218–223
Knorr U, Vinberg M, Kessing LV, Wetterslev J (2010) Salivary cortisol in depressed patients versus control persons: a systematic review and meta-analysis. Psychoneuroendocrinology 35:1275–1286
O'Connor DB, Green JA, Ferguson E, O'Carroll RE, O'Connor RC (2017) Cortisol reactivity and suicidal behavior: Investigating the role of hypothalamic-pituitary-adrenal axis responses to stress in suicide attempters and ideators. Psychoneuroendocrinology 75:183–191
Keilp JG, Stanley BH, Beers SR, Melhem NM, Burke AK, Cooper TB, Oquendo MA, Brent DA, John Mann J (2016) Further evidence of low baseline cortisol levels in suicide attempters. J Affect Disord 190:187–192
Klopfenstein BJ, Purnell JQ, Brandon DD, Isabelle LM, DeBarber AE (2011) Determination of cortisol production rates with contemporary liquid chromatography-mass spectrometry to measure cortisol-d(3) dilution after infusion of deuterated tracer. Clin Biochem 44:430–434
Gao W, Xie Q, Jin J, Qiao T, Wang H, Chen L, Deng H, Lu Z (2010) HPLC-FLU detection of cortisol distribution in human hair. Clin Biochem 43:677–682
Appel D, Schmid RD, Dragan CA, Bureik M, Urlacher VB (2005) A fluorimetric assay for cortisol. Anal Bioanal Chem 383:182–186
Carrozza C, Corsello SM, Paragliola RM, Ingraudo F, Palumbo S, Locantore P, Sferrazza A, Pontecorvi A, Zuppi C (2010) Clinical accuracy of midnight salivary cortisol measured by automated electrochemiluminescence immunoassay method in Cushing's syndrome. Ann Clin Biochem 47:228–232
Lippi G, De Vita F, Salvagno GL, Gelati M, Montagnana M, Guidi GC (2009) Measurement of morning saliva cortisol in athletes. Clin Biochem 42:904–906
Yaneva M, Kirilov G, Zacharieva S (2009) Midnight salivary cortisol, measured by highly sensitive electrochemiluminescence immunoassay, for the diagnosis of Cushing’s syndrome. Open Med 4
Small BC, Davis KB (2002) Validation of a time-resolved fluoroimmunoassay for measuring plasma cortisol in channel catfish Ictalurus punctatus. J World Aquacult Soc 33:184–187
Mitchell JS, Lowe TE, Ingram JR (2009) Rapid ultrasensitive measurement of salivary cortisol using nano-linker chemistry coupled with surface plasmon resonance detection. Analyst 134:380–386
Lee H-J, Lee J-H, Moon H-S, Jang I-S, Choi J-S, Yook J-G, Jung H-I (2012) A planar split-ring resonator-based microwave biosensor for label-free detection of biomolecules. Sens Actuators B Chem 169:26–31
Atashbar MZ, Bejcek B, Vijh A, Singamaneni S (2005) QCM biosensor with ultra thin polymer film. Sens Actuators B Chem 107:945–951
Yamaguchi M, Yoshikawa S, Tahara Y, Niwa D, Imai Y, Shetty V (2009) Point-of-use measurement of salivary cortisol levels. In: 2009 IEEE Sensors, pp 343–346
Arya SK, Dey A, Bhansali S (2011) Polyaniline protected gold nanoparticles based mediator and label free electrochemical cortisol biosensor. Biosens Bioelectron 28:166–173
Vabbina PK, Kaushik A, Pokhrel N, Bhansali S, Pala N (2015) Electrochemical cortisol immunosensors based on sonochemically synthesized zinc oxide 1D nanorods and 2D nanoflakes. Biosens Bioelectron 63:124–130
Venugopal M, Arya SK, Chornokur G, Bhansali S (2011) A Realtime and Continuous Assessment of Cortisol in ISF Using Electrochemical Impedance Spectroscopy. Sens Actuators A Phys 172:154–160
Levine A, Zagoory-Sharon O, Feldman R, Lewis JG, Weller A (2007) Measuring cortisol in human psychobiological studies. Physiol Behav 90:43–53
Brossaud J, Ducint D, Gatta B, Molimard M, Tabarin A, Corcuff J (2012) Urinary cortisol metabolites in corticotroph and adrenal tumours. Endocrine Abstracts 29:55
Russell E, Koren G, Rieder M, Van U, Stan MH (2014) The detection of cortisol in human sweat: implications for measurement of cortisol in hair. Ther Drug Monit 36:30–34
VanBruggen MD, Hackney AC, McMurray RG, Ondrak KS (2011) The relationship between serum and salivary cortisol levels in response to different intensities of exercise. Int J Sports Physiol Perform 6:396–407
Caenegem EV, Wierckx K, Fiers T, Segers H, Vandersypt E, Kaufman JM, T'Sjoen G (2011) Salivary cortisol and testosterone : a comparison of salivary sample collection methods in healthy control. Endocrine Abstracts 26:355
Luppa PB, Sokoll LJ, Chan DW (2001) Immunosensors-principles and applications to clinical chemistry. Clinica Chimica Acta 314:1–26
Lien DH, Kang JS, Amani M, Chen K, Tosun M, Wang HP, Roy T, Eggleston MS, Wu MC, Dubey M, Lee SC, He JH, Javey A (2015) Engineering Light Outcoupling in 2D Materials. Nano Lett. 15:1356–1361
Lin YH, Lin SF, Chi YC, Wu CL, Cheng CH, Tseng WH, He JH, Wu CI, Lee CK, Lin GR (2015) Using n- and p-Type Bi2Te3 Topological Insulator Nanoparticles To Enable Controlled Femtosecond Mode-Locking of Fiber Lasers. Acs Photonics 2:481–490
Cheng B, Li TY, Wei PC, Yin J, Ho KT, Retamal JRD, Mohammed OF, He JH (2018) Layer-edge device of two-dimensional hybrid perovskites. Nat Commun 9:7
Tsai ML, Li MY, Retamal JRD, Lam KT, Lin YC, Suenaga K, Chen LJ, Liang G, Li LJ, He JH (2017) Single atomically sharp lateral monolayer p-n heterojunction solar cells with extraordinarily high power conversion efficiency. Adv Mater 29:7
Gao N, Fang XS (2015) Synthesis and development of graphene inorganic semiconductor nanocomposites. Chem Rev 115:8294–8343
Liu SX, Zheng LX, Yu PP, Han SC, Fang XS (2016) Novel composites of alpha-Fe2O3 tetrakaidecahedron and graphene oxide as an effective photoelectrode with enhanced photocurrent performances. Adv Funct Mater. 26:3331–3339
Lee CP, Lai KY, Lin CA, Li CT, Ho KC, Wu CI, Lau SP, He JH (2017) A paper-based electrode using a graphene dot/PEDOT:PSS composite for flexible solar cells. Nano Energy 36:260–267
Ouyang WX, Teng F, Fang XS (2018) High performance BiOCl nanosheets/TiO2 nanotube arrays heterojunction UV photodetector: the influences of self-induced inner electric fields in the BiOCl nanosheets. Adv Funct Mater. 28:12
Deshpande NG, Sagade AA, Gudage YG, Lokhande CD, Sharma R (2007) Growth and characterization of tin disulfide (SnS2) thin film deposited by successive ionic layer adsorption and reaction (SILAR) technique. J Alloys Compounds 436:421–426
Panda SK, Antonakos A, Liarokapis E, Bhattacharya S, Chaudhuri S (2007) Optical properties of nanocrystalline SnS2 thin films. Materials Res Bull 42:576–583
Wang C, Tang K, Yang Q, Qian Y (2002) Raman scattering, far infrared spectrum and photoluminescence of SnS2 nanocrystallites. Chem Phys Lett 357:371–375
Yang YB, Dash JK, Littlejohn AJ, Xiang Y, Wang Y, Shi J, Zhang LH, Kisslinger K, Lu TM, Wang GC (2016) Large single crystal SnS2 flakes synthesized from coevaporation of Sn and S. Crystal Growth Design 16:961–973
Chen X, Hou Y, Zhang B, Yang XH, Yang HG (2013) Low-cost SnS(x) counter electrodes for dye-sensitized solar cells. Chem Commun (Camb) 49:5793–5795
Yang B, Zuo X, Chen P, Zhou L, Yang X, Zhang H, Li G, Wu M, Ma Y, Jin S, Chen X (2015) Nanocomposite of tin sulfide nanoparticles with reduced graphene oxide in high-efficiency dye-sensitized solar cells. ACS Appl Mater Interfaces 7:137–143
Youn DH, Stauffer SK, Xiao P, Park H, Nam Y, Dolocan A, Henkelman G, Heller A, Mullins CB (2016) Simple synthesis of nanocrystalline tin sulfide/N-doped reduced graphene oxide composites as lithium ion battery anodes. ACS Nano 10:10778–10788
Gao C, Li L, Raji AR, Kovalchuk A, Peng Z, Fei H, He Y, Kim ND, Zhong Q, Xie E, Tour JM (2015) Tin disulfide nanoplates on graphene nanoribbons for full lithium ion batteries. ACS Appl Mater Interfaces 7:26549–26556
Yang Z, Ren Y, Zhang Y, Li J, Li H, Hu X, Xu Q (2011) Nanoflake-like SnS2 matrix for glucose biosensing based on direct electrochemistry of glucose oxidase. Biosens Bioelectron 26:4337–4341
Li Y, Leonardi SG, Bonavita A, Neri G, Wlodarski W (2016) Two-dimensional (2D) SnS2-based oxygen sensor. Procedia Engineering 168:1102–1105
Moreno-Guzman M, Eguilaz M, Campuzano S, Gonzalez-Cortes A, Yanez-Sedeno P, Pingarron JM (2010) Disposable immunosensor for cortisol using functionalized magnetic particles. Analyst 135:1926–1933
Yamaguchi M, Matsuda Y, Yoshikawa S, Sasaki M, Imai Y, Niwa D, Shetty V (2011) Rapid hormone immunosensor with fluid control mechanism. In: 2011 16th International Solid-State Sensors, Actuators and Microsystems Conference, pp 1164–1167
Sun B, Gou Y, Ma Y, Zheng X, Bai R, Ahmed Abdelmoaty AA, Hu F (2017) Investigate electrochemical immunosensor of cortisol based on gold nanoparticles/magnetic functionalized reduced graphene oxide. Biosensors and Bioelectronics 88:55–62
Arya SK, Chornokur G, Venugopal M, Bhansali S (2010) Antibody modified gold micro array electrode based electrochemical immunosensor for ultrasensitive detection of cortisol in saliva and ISF. Procedia Engineering 5:804–807
Arya SK, Chornokur G, Venugopal M, Bhansali S (2010) Antibody functionalized interdigitated [small mu ]-electrode (ID[small mu ]E) based impedimetric cortisol biosensor. Analyst 135:1941–1946
Liu X, Zhao R, Mao W, Feng H, Liu X, Wong DKY (2011) Detection of cortisol at a gold nanoparticle|Protein G–DTBP-scaffold modified electrochemical immunosensor. The Analyst 136:5204
Ueno Y, Furukawa K, Hayashi K, Takamura M, Hibino H, Tamechika E (2013) Graphene-modified Interdigitated Array Electrode: Fabrication, Characterization, and Electrochemical Immunoassay Application. Anal Sci 29:55–60
Kaushik A, Vasudev A, Arya SK, Bhansali S (2013) Mediator and label free estimation of stress biomarker using electrophoretically deposited Ag@AgO–polyaniline hybrid nanocomposite. Biosens Bioelectron 50:35–41
We thank the reviewers for their valuable comments.
The authors are grateful for the financial support for this research by the Ministry of Science and Technology of Taiwan (MOST 104-2622-E-027-027-CC3; MOST 104-2221-E-027-061, MOST 105-2221-E-027-028, MOST 106-2221-E-027-034), the National Taipei University of Technology–Shenzhen University Joint Research Program (NTUT-SZU-107-01 (2018001); NTUT-SZU-108-05 (2019005)); in part from the National Natural Science Foundation of China (61504083), Guangdong Province Key Research and Development Plan (2019B010138002), and partly from the Development and Reform Commission of Jilin Province, under grant no. 2017C059-5.
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Liu, X., Hsu, S.P.C., Liu, W. et al. Salivary Electrochemical Cortisol Biosensor Based on Tin Disulfide Nanoflakes. Nanoscale Res Lett 14, 189 (2019). https://doi.org/10.1186/s11671-019-3012-0
- 2D Tin disulfide nanoflakes
- Electrochemical biosensor
- Enzyme-linked immunosorbent assay